-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171028_095852_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.047277 to 34.183139
-each separately from 0s to 34s and from 47277us to 563110us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.515255 in 1 packets - #/No: 18/18
      15.619740 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-28T09:58:55.047.277.000 to 2017-10-28T09:59:29.183.139.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.515255 in 1 packets - #/No: 18/18
      15.619740 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4334712 to 35.4333382
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999296 in 1 packets - #/No: 18/18
      15.9999374 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171028_095852_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4807487 to 1.4810377
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5151830  -- OK for SL --
   MIU time from 2017-10-28T09:58:55.047.277.000 to 2017-10-28T09:58:55.047.566.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.515255

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.9962207 to 19.9965097
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6196790  -- OK for SL --
   MIU time from 2017-10-28T09:59:13.562.821.000 to 2017-10-28T09:59:13.563.110.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.619740

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.6161887 to 35.6164777
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-28T09:59:29.182.850.000 to 2017-10-28T09:59:29.183.139.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-28T09:58:55.047.136.750
   Max SL corrected packet time: 2017-10-28T09:59:29.182.998.750
   Min SL corrected XO rel time: 0.0472777
   Max SL corrected XO rel time: 34.1831402
   TM values from -20 to 9 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-28T09:58:55.047.136.750 (XO: 0.0472777, t=34.0°C)  Rank: 00003
   TM values: Max =     9  Min =   -19  Ave =    -4.41  Std =     5.39
   Min TM value:-19 at an offset of 168.888 usec -> 2017-10-28T09:58:55.047.305.638 (XO: 0.0474466)
   Max TM value:9 at an offset of 189.413 usec -> 2017-10-28T09:58:55.047.326.163 (XO: 0.0474671)
   Max dTM/dt [1/us]::1.06e+02 at an offset of 234.963 usec -> 2017-10-28T09:58:55.047.371.713 (XO: 0.0475126)
   Min dTM/dt [1/us]::-1.18e+02 at an offset of 189.726 usec -> 2017-10-28T09:58:55.047.326.475 (XO: 0.0474674)

 - Block number: 2 starting at 2017-10-28T09:59:13.562.680.750 (XO: 18.5628221, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -16  Ave =    -4.35  Std =     3.40
   Min TM value:-16 at an offset of 102.525 usec -> 2017-10-28T09:59:13.562.783.275 (XO: 18.5629246)
   Max TM value:7 at an offset of 235.538 usec -> 2017-10-28T09:59:13.562.916.288 (XO: 18.5630576)
   Max dTM/dt [1/us]::9.10e+01 at an offset of 124.188 usec -> 2017-10-28T09:59:13.562.804.937 (XO: 18.5629463)
   Min dTM/dt [1/us]::-8.90e+01 at an offset of 220.863 usec -> 2017-10-28T09:59:13.562.901.613 (XO: 18.5630430)

 - Block number: 3 starting at 2017-10-28T09:59:29.182.709.750 (XO: 34.1828512, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -20  Ave =    -4.38  Std =     4.09
   Min TM value:-20 at an offset of 82.375 usec -> 2017-10-28T09:59:29.182.792.125 (XO: 34.1829336)
   Max TM value:9 at an offset of 52.863 usec -> 2017-10-28T09:59:29.182.762.612 (XO: 34.1829041)
   Max dTM/dt [1/us]::1.25e+02 at an offset of 86.125 usec -> 2017-10-28T09:59:29.182.795.875 (XO: 34.1829373)
   Min dTM/dt [1/us]::-1.19e+02 at an offset of 120.125 usec -> 2017-10-28T09:59:29.182.829.875 (XO: 34.1829713)
