-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171027_101620_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.095083 to 30.120065
-each separately from 0s to 30s and from 95083us to 288270us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      25.192609 in 1 packets - #/No: 18/18
      4.831506 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-27T10:16:23.095.083.000 to 2017-10-27T10:16:53.120.065.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      25.192609 in 1 packets - #/No: 18/18
      4.831506 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9391727 to 31.9390552
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.9999021 in 1 packets - #/No: 18/18
      4.9999804 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171027_101620_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0342564 to 2.0345454
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.1925110  -- OK for SL --
   MIU time from 2017-10-27T10:16:23.095.083.000 to 2017-10-27T10:16:23.095.372.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.192609

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.2270564 to 27.2273454
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.8314870  -- OK for SL --
   MIU time from 2017-10-27T10:16:48.287.981.000 to 2017-10-27T10:16:48.288.270.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.831506

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.0588324 to 32.0591214
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-27T10:16:53.119.776.000 to 2017-10-27T10:16:53.120.065.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-27T10:16:23.094.942.750
   Max SL corrected packet time: 2017-10-27T10:16:53.119.924.750
   Min SL corrected XO rel time: 0.0950841
   Max SL corrected XO rel time: 30.1200667
   TM values from -24 to 16 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-27T10:16:23.094.942.750 (XO: 0.0950841, t=35.6°C)  Rank: 00002
   TM values: Max =     7  Min =   -15  Ave =    -4.42  Std =     3.10
   Min TM value:-15 at an offset of 109.625 usec -> 2017-10-27T10:16:23.095.052.375 (XO: 0.0951937)
   Max TM value:7 at an offset of 195.776 usec -> 2017-10-27T10:16:23.095.138.525 (XO: 0.0952798)
   Max dTM/dt [1/us]::7.70e+01 at an offset of 128.376 usec -> 2017-10-27T10:16:23.095.071.125 (XO: 0.0952124)
   Min dTM/dt [1/us]::-6.70e+01 at an offset of 95.425 usec -> 2017-10-27T10:16:23.095.038.175 (XO: 0.0951795)

 - Block number: 2 starting at 2017-10-27T10:16:48.287.840.750 (XO: 25.2879827, t=35.6°C)  Rank: 00006
   TM values: Max =    16  Min =   -24  Ave =    -4.31  Std =     5.48
   Min TM value:-24 at an offset of 112.150 usec -> 2017-10-27T10:16:48.287.952.900 (XO: 25.2880949)
   Max TM value:16 at an offset of 143.451 usec -> 2017-10-27T10:16:48.287.984.200 (XO: 25.2881262)
   Max dTM/dt [1/us]::1.44e+02 at an offset of 119.213 usec -> 2017-10-27T10:16:48.287.959.962 (XO: 25.2881019)
   Min dTM/dt [1/us]::-1.37e+02 at an offset of 99.813 usec -> 2017-10-27T10:16:48.287.940.562 (XO: 25.2880825)

 - Block number: 3 starting at 2017-10-27T10:16:53.119.635.750 (XO: 30.1197777, t=35.6°C)  Rank: 00005
   TM values: Max =    11  Min =   -20  Ave =    -4.37  Std =     5.32
   Min TM value:-20 at an offset of 108.713 usec -> 2017-10-27T10:16:53.119.744.462 (XO: 30.1198864)
   Max TM value:11 at an offset of 138.488 usec -> 2017-10-27T10:16:53.119.774.238 (XO: 30.1199162)
   Max dTM/dt [1/us]::1.23e+02 at an offset of 138.901 usec -> 2017-10-27T10:16:53.119.774.650 (XO: 30.1199166)
   Min dTM/dt [1/us]::-1.19e+02 at an offset of 116.063 usec -> 2017-10-27T10:16:53.119.751.812 (XO: 30.1198937)
