-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171027_054356_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 9.033292 to 36.534280
-each separately from 9s to 36s and from 13155us to 534280us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.979574 in 1 packets - #/No: 18/18
      21.520547 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-27T05:44:08.033.292.000 to 2017-10-27T05:44:35.534.280.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.979574 in 1 packets - #/No: 18/18
      21.520547 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4409295 to 37.4408238
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.9999765 in 1 packets - #/No: 18/18
      20.9999178 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171027_054356_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.4742224 to 10.4745114
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.9795510  -- OK for SL --
   MIU time from 2017-10-27T05:44:08.033.292.000 to 2017-10-27T05:44:08.033.581.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.979574

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4540624 to 16.4543514
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.5204630  -- OK for SL --
   MIU time from 2017-10-27T05:44:14.013.155.000 to 2017-10-27T05:44:14.013.444.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.520547

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.9748144 to 37.9751034
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-27T05:44:35.533.991.000 to 2017-10-27T05:44:35.534.280.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-27T05:44:08.033.151.750
   Max SL corrected packet time: 2017-10-27T05:44:35.534.139.750
   Min SL corrected XO rel time: 9.0332930
   Max SL corrected XO rel time: 36.5342817
   TM values from -23 to 11 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-27T05:44:08.033.151.750 (XO: 9.0332930, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -23  Ave =    -4.44  Std =     4.89
   Min TM value:-23 at an offset of 154.226 usec -> 2017-10-27T05:44:08.033.305.975 (XO: 9.0334473)
   Max TM value:11 at an offset of 114.175 usec -> 2017-10-27T05:44:08.033.265.925 (XO: 9.0334072)
   Max dTM/dt [1/us]::2.62e+02 at an offset of 107.300 usec -> 2017-10-27T05:44:08.033.259.050 (XO: 9.0334003)
   Min dTM/dt [1/us]::-2.79e+02 at an offset of 154.188 usec -> 2017-10-27T05:44:08.033.305.938 (XO: 9.0334472)

 - Block number: 2 starting at 2017-10-27T05:44:14.013.014.750 (XO: 15.0131565, t=34.0°C)  Rank: 00002
   TM values: Max =    11  Min =   -19  Ave =    -4.48  Std =     4.52
   Min TM value:-19 at an offset of 251.101 usec -> 2017-10-27T05:44:14.013.265.850 (XO: 15.0134076)
   Max TM value:11 at an offset of 302.351 usec -> 2017-10-27T05:44:14.013.317.101 (XO: 15.0134588)
   Max dTM/dt [1/us]::2.05e+02 at an offset of 268.526 usec -> 2017-10-27T05:44:14.013.283.276 (XO: 15.0134250)
   Min dTM/dt [1/us]::-2.00e+02 at an offset of 222.163 usec -> 2017-10-27T05:44:14.013.236.913 (XO: 15.0133786)

 - Block number: 3 starting at 2017-10-27T05:44:35.533.850.750 (XO: 36.5339927, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -20  Ave =    -4.45  Std =     4.85
   Min TM value:-20 at an offset of 118.500 usec -> 2017-10-27T05:44:35.533.969.250 (XO: 36.5341112)
   Max TM value:11 at an offset of 116.125 usec -> 2017-10-27T05:44:35.533.966.875 (XO: 36.5341088)
   Max dTM/dt [1/us]::2.11e+02 at an offset of 130.726 usec -> 2017-10-27T05:44:35.533.981.475 (XO: 36.5341234)
   Min dTM/dt [1/us]::-1.96e+02 at an offset of 89.363 usec -> 2017-10-27T05:44:35.533.940.112 (XO: 36.5340820)
