-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171025_215708_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.277384 to 32.688067
-each separately from 8s to 32s and from 277384us to 688067us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.266362 in 1 packets - #/No: 18/18
      17.143454 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-25T21:57:19.277.384.000 to 2017-10-25T21:57:43.688.067.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.266362 in 1 packets - #/No: 18/18
      17.143454 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4370582 to 33.4369692
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999741 in 1 packets - #/No: 18/18
      16.9999369 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171025_215708_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.7144424 to 9.7147314
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2663350  -- OK for SL --
   MIU time from 2017-10-25T21:57:19.277.384.000 to 2017-10-25T21:57:19.277.673.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.266362

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9810664 to 16.9813554
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1433910  -- OK for SL --
   MIU time from 2017-10-25T21:57:26.544.035.000 to 2017-10-25T21:57:26.544.324.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.143454

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.1247464 to 34.1250354
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-25T21:57:43.687.778.000 to 2017-10-25T21:57:43.688.067.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-25T21:57:19.277.243.750
   Max SL corrected packet time: 2017-10-25T21:57:43.687.926.750
   Min SL corrected XO rel time: 8.2773852
   Max SL corrected XO rel time: 32.6880687
   TM values from -27 to 23 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-25T21:57:19.277.243.750 (XO: 8.2773852, t=34.0°C)  Rank: 00004
   TM values: Max =    23  Min =   -27  Ave =    -4.43  Std =     4.79
   Min TM value:-27 at an offset of 85.338 usec -> 2017-10-25T21:57:19.277.329.087 (XO: 8.2774706)
   Max TM value:23 at an offset of 86.063 usec -> 2017-10-25T21:57:19.277.329.812 (XO: 8.2774713)
   Max dTM/dt [1/us]::2.00e+02 at an offset of 85.888 usec -> 2017-10-25T21:57:19.277.329.637 (XO: 8.2774711)
   Min dTM/dt [1/us]::-2.02e+02 at an offset of 14.550 usec -> 2017-10-25T21:57:19.277.258.300 (XO: 8.2773998)

 - Block number: 2 starting at 2017-10-25T21:57:26.543.894.750 (XO: 15.5440361, t=34.0°C)  Rank: 00004
   TM values: Max =    14  Min =   -22  Ave =    -4.39  Std =     4.79
   Min TM value:-22 at an offset of 71.838 usec -> 2017-10-25T21:57:26.543.966.587 (XO: 15.5441080)
   Max TM value:14 at an offset of 265.176 usec -> 2017-10-25T21:57:26.544.159.925 (XO: 15.5443013)
   Max dTM/dt [1/us]::2.43e+02 at an offset of 168.038 usec -> 2017-10-25T21:57:26.544.062.788 (XO: 15.5442042)
   Min dTM/dt [1/us]::-2.56e+02 at an offset of 115.513 usec -> 2017-10-25T21:57:26.544.010.262 (XO: 15.5441516)

 - Block number: 3 starting at 2017-10-25T21:57:43.687.637.750 (XO: 32.6877797, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -24  Ave =    -4.40  Std =     4.53
   Min TM value:-24 at an offset of 285.464 usec -> 2017-10-25T21:57:43.687.923.213 (XO: 32.6880652)
   Max TM value:13 at an offset of 75.250 usec -> 2017-10-25T21:57:43.687.713.000 (XO: 32.6878550)
   Max dTM/dt [1/us]::2.19e+02 at an offset of 8.213 usec -> 2017-10-25T21:57:43.687.645.962 (XO: 32.6877880)
   Min dTM/dt [1/us]::-1.95e+02 at an offset of 198.301 usec -> 2017-10-25T21:57:43.687.836.050 (XO: 32.6879780)
