-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171025_172444_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.353132 to 30.192738
-each separately from 0s to 30s and from 192449us to 458504us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.104794 in 1 packets - #/No: 18/18
      10.733945 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-25T17:24:47.353.132.000 to 2017-10-25T17:25:17.192.738.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.104794 in 1 packets - #/No: 18/18
      10.733945 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4374532 to 31.4373353
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.9999253 in 1 packets - #/No: 18/18
      10.9999568 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171025_172444_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7905852 to 1.7908742
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1047190  -- OK for SL --
   MIU time from 2017-10-25T17:24:47.353.132.000 to 2017-10-25T17:24:47.353.421.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.104794

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.8955932 to 20.8958822
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7339030  -- OK for SL --
   MIU time from 2017-10-25T17:25:06.458.215.000 to 2017-10-25T17:25:06.458.504.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.733945

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.6297852 to 31.6300742
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-25T17:25:17.192.449.000 to 2017-10-25T17:25:17.192.738.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-25T17:24:47.352.991.750
   Max SL corrected packet time: 2017-10-25T17:25:17.192.597.750
   Min SL corrected XO rel time: 0.3531334
   Max SL corrected XO rel time: 30.1927397
   TM values from -33 to 23 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-25T17:24:47.352.991.750 (XO: 0.3531334, t=34.0°C)  Rank: 00010
   TM values: Max =    23  Min =   -33  Ave =    -4.50  Std =    10.97
   Min TM value:-33 at an offset of 216.601 usec -> 2017-10-25T17:24:47.353.208.350 (XO: 0.3533500)
   Max TM value:23 at an offset of 216.226 usec -> 2017-10-25T17:24:47.353.207.975 (XO: 0.3533496)
   Max dTM/dt [1/us]::1.17e+02 at an offset of 305.964 usec -> 2017-10-25T17:24:47.353.297.713 (XO: 0.3534393)
   Min dTM/dt [1/us]::-1.19e+02 at an offset of 228.713 usec -> 2017-10-25T17:24:47.353.220.463 (XO: 0.3533621)

 - Block number: 2 starting at 2017-10-25T17:25:06.458.074.750 (XO: 19.4582165, t=34.0°C)  Rank: 00007
   TM values: Max =    20  Min =   -26  Ave =    -4.45  Std =     7.55
   Min TM value:-26 at an offset of 123.438 usec -> 2017-10-25T17:25:06.458.198.187 (XO: 19.4583399)
   Max TM value:20 at an offset of 136.376 usec -> 2017-10-25T17:25:06.458.211.125 (XO: 19.4583529)
   Max dTM/dt [1/us]::1.23e+02 at an offset of 123.038 usec -> 2017-10-25T17:25:06.458.197.787 (XO: 19.4583395)
   Min dTM/dt [1/us]::-1.19e+02 at an offset of 218.013 usec -> 2017-10-25T17:25:06.458.292.763 (XO: 19.4584345)

 - Block number: 3 starting at 2017-10-25T17:25:17.192.308.750 (XO: 30.1924507, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -20  Ave =    -4.36  Std =     4.76
   Min TM value:-20 at an offset of 73.525 usec -> 2017-10-25T17:25:17.192.382.275 (XO: 30.1925242)
   Max TM value:12 at an offset of 232.138 usec -> 2017-10-25T17:25:17.192.540.888 (XO: 30.1926828)
   Max dTM/dt [1/us]::1.63e+02 at an offset of 197.126 usec -> 2017-10-25T17:25:17.192.505.875 (XO: 30.1926478)
   Min dTM/dt [1/us]::-1.63e+02 at an offset of 120.338 usec -> 2017-10-25T17:25:17.192.429.087 (XO: 30.1925710)
