-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171010_100152_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.759318 to 30.022286
-each separately from 11s to 30s and from 10995us to 759607us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.251388 in 1 packets - #/No: 18/18
      15.010713 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-10T10:02:06.759.318.000 to 2017-10-10T10:02:25.022.286.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.251388 in 1 packets - #/No: 18/18
      15.010713 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.4391476 to 31.4390740
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.9999845 in 1 packets - #/No: 18/18
      14.9999419 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171010_100152_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.1984639 to 13.1987529
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.2513750  -- OK for SL --
   MIU time from 2017-10-10T10:02:06.759.318.000 to 2017-10-10T10:02:06.759.607.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.251388

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4501279 to 16.4504169
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2017-10-10T10:02:10.010.995.000 to 2017-10-10T10:02:10.011.284.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010713

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4610719 to 31.4613609
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-10T10:02:25.021.997.000 to 2017-10-10T10:02:25.022.286.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-10T10:02:06.759.177.750
   Max SL corrected packet time: 2017-10-10T10:02:25.022.145.750
   Min SL corrected XO rel time: 11.7593193
   Max SL corrected XO rel time: 30.0222870
   TM values from -13 to 4 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-10T10:02:06.759.177.750 (XO: 11.7593193, t=34.0°C)  Rank: 00002
   TM values: Max =     3  Min =   -12  Ave =    -4.49  Std =     2.05
   Min TM value:-12 at an offset of 22.975 usec -> 2017-10-10T10:02:06.759.200.725 (XO: 11.7593422)
   Max TM value:3 at an offset of 16.100 usec -> 2017-10-10T10:02:06.759.193.850 (XO: 11.7593354)
   Max dTM/dt [1/us]::6.10e+01 at an offset of 256.213 usec -> 2017-10-10T10:02:06.759.433.963 (XO: 11.7595755)
   Min dTM/dt [1/us]::-6.20e+01 at an offset of 296.851 usec -> 2017-10-10T10:02:06.759.474.601 (XO: 11.7596161)

 - Block number: 2 starting at 2017-10-10T10:02:10.010.854.750 (XO: 15.0109958, t=34.0°C)  Rank: 00001
   TM values: Max =     4  Min =   -13  Ave =    -4.54  Std =     2.37
   Min TM value:-13 at an offset of 239.563 usec -> 2017-10-10T10:02:10.011.094.313 (XO: 15.0112354)
   Max TM value:4 at an offset of 25.988 usec -> 2017-10-10T10:02:10.010.880.737 (XO: 15.0110218)
   Max dTM/dt [1/us]::7.00e+01 at an offset of 264.476 usec -> 2017-10-10T10:02:10.011.119.226 (XO: 15.0112603)
   Min dTM/dt [1/us]::-6.70e+01 at an offset of 90.238 usec -> 2017-10-10T10:02:10.010.944.987 (XO: 15.0110861)

 - Block number: 3 starting at 2017-10-10T10:02:25.021.856.750 (XO: 30.0219980, t=34.0°C)  Rank: 00001
   TM values: Max =     3  Min =   -11  Ave =    -4.46  Std =     1.85
   Min TM value:-11 at an offset of 13.950 usec -> 2017-10-10T10:02:25.021.870.700 (XO: 30.0220119)
   Max TM value:3 at an offset of 130.376 usec -> 2017-10-10T10:02:25.021.987.125 (XO: 30.0221284)
   Max dTM/dt [1/us]::6.20e+01 at an offset of 136.076 usec -> 2017-10-10T10:02:25.021.992.825 (XO: 30.0221341)
   Min dTM/dt [1/us]::-6.60e+01 at an offset of 93.875 usec -> 2017-10-10T10:02:25.021.950.625 (XO: 30.0220919)
