-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171008_140840_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.165143 to 35.543756
-each separately from 10s to 35s and from 11531us to 543756us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.846099 in 1 packets - #/No: 18/18
      20.531647 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-08T14:08:53.165.143.000 to 2017-10-08T14:09:18.543.756.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.846099 in 1 packets - #/No: 18/18
      20.531647 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4376276 to 36.4375295
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.9999803 in 1 packets - #/No: 18/18
      19.9999216 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      C0 -> 35.6°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171008_140840_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.6027716 to 11.6030606
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.8460790  -- OK for SL --
   MIU time from 2017-10-08T14:08:53.165.143.000 to 2017-10-08T14:08:53.165.432.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.846099

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4491396 to 16.4494286
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5315670  -- OK for SL --
   MIU time from 2017-10-08T14:08:58.011.531.000 to 2017-10-08T14:08:58.011.820.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.531647

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.9809956 to 36.9812846
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-08T14:09:18.543.467.000 to 2017-10-08T14:09:18.543.756.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-08T14:08:53.165.002.750
   Max SL corrected packet time: 2017-10-08T14:09:18.543.615.750
   Min SL corrected XO rel time: 10.1651446
   Max SL corrected XO rel time: 35.5437573
   TM values from -25 to 15 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-08T14:08:53.165.002.750 (XO: 10.1651446, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -24  Ave =    -4.54  Std =     5.80
   Min TM value:-24 at an offset of 105.100 usec -> 2017-10-08T14:08:53.165.107.850 (XO: 10.1652497)
   Max TM value:13 at an offset of 149.788 usec -> 2017-10-08T14:08:53.165.152.538 (XO: 10.1652944)
   Max dTM/dt [1/us]::1.48e+02 at an offset of 131.438 usec -> 2017-10-08T14:08:53.165.134.188 (XO: 10.1652761)
   Min dTM/dt [1/us]::-1.38e+02 at an offset of 115.350 usec -> 2017-10-08T14:08:53.165.118.100 (XO: 10.1652600)

 - Block number: 2 starting at 2017-10-08T14:08:58.011.390.750 (XO: 15.0115317, t=34.0°C)  Rank: 00002
   TM values: Max =    11  Min =   -21  Ave =    -4.65  Std =     4.89
   Min TM value:-21 at an offset of 34.425 usec -> 2017-10-08T14:08:58.011.425.175 (XO: 15.0115662)
   Max TM value:11 at an offset of 112.800 usec -> 2017-10-08T14:08:58.011.503.550 (XO: 15.0116445)
   Max dTM/dt [1/us]::1.30e+02 at an offset of 24.350 usec -> 2017-10-08T14:08:58.011.415.100 (XO: 15.0115561)
   Min dTM/dt [1/us]::-1.35e+02 at an offset of 28.900 usec -> 2017-10-08T14:08:58.011.419.650 (XO: 15.0115606)

 - Block number: 3 starting at 2017-10-08T14:09:18.543.326.750 (XO: 35.5434683, t=34.0°C)  Rank: 00005
   TM values: Max =    15  Min =   -25  Ave =    -4.58  Std =     6.13
   Min TM value:-25 at an offset of 187.863 usec -> 2017-10-08T14:09:18.543.514.613 (XO: 35.5436561)
   Max TM value:15 at an offset of 121.525 usec -> 2017-10-08T14:09:18.543.448.275 (XO: 35.5435898)
   Max dTM/dt [1/us]::1.47e+02 at an offset of 127.013 usec -> 2017-10-08T14:09:18.543.453.763 (XO: 35.5435953)
   Min dTM/dt [1/us]::-1.36e+02 at an offset of 112.813 usec -> 2017-10-08T14:09:18.543.439.562 (XO: 35.5435811)
