-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171008_083544_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.850587 to 30.158711
-each separately from 2s to 30s and from 158422us to 862598us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.011433 in 1 packets - #/No: 18/18
      12.295824 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-08T08:35:49.850.587.000 to 2017-10-08T08:36:17.158.711.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.011433 in 1 packets - #/No: 18/18
      12.295824 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4322176 to 31.4321079
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999413 in 1 packets - #/No: 18/18
      12.9999490 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171008_083544_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2828023 to 4.2830913
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0113750  -- OK for SL --
   MIU time from 2017-10-08T08:35:49.850.587.000 to 2017-10-08T08:35:49.850.876.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.011433

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.2944663 to 19.2947553
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2957750  -- OK for SL --
   MIU time from 2017-10-08T08:36:04.862.309.000 to 2017-10-08T08:36:04.862.598.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.295824

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5905303 to 31.5908193
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-08T08:36:17.158.422.000 to 2017-10-08T08:36:17.158.711.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-08T08:35:49.850.446.750
   Max SL corrected packet time: 2017-10-08T08:36:17.158.570.750
   Min SL corrected XO rel time: 2.8505880
   Max SL corrected XO rel time: 30.1587120
   TM values from -17 to 9 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-08T08:35:49.850.446.750 (XO: 2.8505880, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -16  Ave =    -4.35  Std =     3.43
   Min TM value:-16 at an offset of 200.463 usec -> 2017-10-08T08:35:49.850.647.213 (XO: 2.8507885)
   Max TM value:7 at an offset of 172.413 usec -> 2017-10-08T08:35:49.850.619.163 (XO: 2.8507604)
   Max dTM/dt [1/us]::9.30e+01 at an offset of 222.101 usec -> 2017-10-08T08:35:49.850.668.850 (XO: 2.8508101)
   Min dTM/dt [1/us]::-7.50e+01 at an offset of 100.025 usec -> 2017-10-08T08:35:49.850.546.775 (XO: 2.8506880)

 - Block number: 2 starting at 2017-10-08T08:36:04.862.168.750 (XO: 17.8623108, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -15  Ave =    -4.40  Std =     2.82
   Min TM value:-15 at an offset of 47.075 usec -> 2017-10-08T08:36:04.862.215.825 (XO: 17.8623578)
   Max TM value:6 at an offset of 46.250 usec -> 2017-10-08T08:36:04.862.215.000 (XO: 17.8623570)
   Max dTM/dt [1/us]::7.90e+01 at an offset of 58.000 usec -> 2017-10-08T08:36:04.862.226.750 (XO: 17.8623688)
   Min dTM/dt [1/us]::-7.60e+01 at an offset of 259.714 usec -> 2017-10-08T08:36:04.862.428.463 (XO: 17.8625705)

 - Block number: 3 starting at 2017-10-08T08:36:17.158.281.750 (XO: 30.1584230, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -17  Ave =    -4.41  Std =     4.02
   Min TM value:-17 at an offset of 99.363 usec -> 2017-10-08T08:36:17.158.381.112 (XO: 30.1585224)
   Max TM value:9 at an offset of 164.113 usec -> 2017-10-08T08:36:17.158.445.863 (XO: 30.1585871)
   Max dTM/dt [1/us]::1.00e+02 at an offset of 305.989 usec -> 2017-10-08T08:36:17.158.587.738 (XO: 30.1587290)
   Min dTM/dt [1/us]::-7.80e+01 at an offset of 112.513 usec -> 2017-10-08T08:36:17.158.394.262 (XO: 30.1585355)
