-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171002_162309_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.663407 to 33.597343
-each separately from 8s to 33s and from 109916us to 663696us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.446220 in 1 packets - #/No: 18/18
      17.486849 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-02T16:23:20.663.407.000 to 2017-10-02T16:23:45.597.343.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.446220 in 1 packets - #/No: 18/18
      17.486849 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9362102 to 34.9361153
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999696 in 1 packets - #/No: 18/18
      16.9999355 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171002_162309_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5996164 to 10.5999054
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.4461910  -- OK for SL --
   MIU time from 2017-10-02T16:23:20.663.407.000 to 2017-10-02T16:23:20.663.696.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.446220

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.0460964 to 18.0463854
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4867830  -- OK for SL --
   MIU time from 2017-10-02T16:23:28.109.916.000 to 2017-10-02T16:23:28.110.205.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.486849

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.5331684 to 35.5334574
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-02T16:23:45.597.054.000 to 2017-10-02T16:23:45.597.343.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-02T16:23:20.663.266.750
   Max SL corrected packet time: 2017-10-02T16:23:45.597.202.750
   Min SL corrected XO rel time: 8.6634087
   Max SL corrected XO rel time: 33.5973444
   TM values from -36 to 27 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-02T16:23:20.663.266.750 (XO: 8.6634087, t=35.6°C)  Rank: 00011
   TM values: Max =    27  Min =   -36  Ave =    -4.50  Std =     9.21
   Min TM value:-36 at an offset of 134.263 usec -> 2017-10-02T16:23:20.663.401.013 (XO: 8.6635430)
   Max TM value:27 at an offset of 116.088 usec -> 2017-10-02T16:23:20.663.382.837 (XO: 8.6635248)
   Max dTM/dt [1/us]::2.90e+02 at an offset of 116.363 usec -> 2017-10-02T16:23:20.663.383.112 (XO: 8.6635251)
   Min dTM/dt [1/us]::-2.94e+02 at an offset of 134.213 usec -> 2017-10-02T16:23:20.663.400.963 (XO: 8.6635429)

 - Block number: 2 starting at 2017-10-02T16:23:28.109.775.750 (XO: 16.1099170, t=35.6°C)  Rank: 00006
   TM values: Max =    21  Min =   -28  Ave =    -4.36  Std =     7.90
   Min TM value:-28 at an offset of 108.213 usec -> 2017-10-02T16:23:28.109.883.962 (XO: 16.1100252)
   Max TM value:21 at an offset of 133.763 usec -> 2017-10-02T16:23:28.109.909.513 (XO: 16.1100508)
   Max dTM/dt [1/us]::2.41e+02 at an offset of 164.888 usec -> 2017-10-02T16:23:28.109.940.638 (XO: 16.1100819)
   Min dTM/dt [1/us]::-2.68e+02 at an offset of 255.751 usec -> 2017-10-02T16:23:28.110.031.500 (XO: 16.1101728)

 - Block number: 3 starting at 2017-10-02T16:23:45.596.913.750 (XO: 33.5970554, t=35.6°C)  Rank: 00006
   TM values: Max =    21  Min =   -28  Ave =    -4.46  Std =     7.21
   Min TM value:-28 at an offset of 108.788 usec -> 2017-10-02T16:23:45.597.022.537 (XO: 33.5971642)
   Max TM value:21 at an offset of 159.676 usec -> 2017-10-02T16:23:45.597.073.425 (XO: 33.5972150)
   Max dTM/dt [1/us]::2.62e+02 at an offset of 129.438 usec -> 2017-10-02T16:23:45.597.043.187 (XO: 33.5971848)
   Min dTM/dt [1/us]::-2.65e+02 at an offset of 125.000 usec -> 2017-10-02T16:23:45.597.038.750 (XO: 33.5971804)
