-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171001_074945_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.320616 to 35.561925
-each separately from 5s to 35s and from 320616us to 561925us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.147016 in 1 packets - #/No: 18/18
      14.093426 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-01T07:49:53.320.616.000 to 2017-10-01T07:50:23.561.925.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.147016 in 1 packets - #/No: 18/18
      14.093426 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.5462636 to 36.5461568
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999431 in 1 packets - #/No: 18/18
      13.9999501 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171001_074945_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.8668797 to 6.8671687
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1469590  -- OK for SL --
   MIU time from 2017-10-01T07:49:53.320.616.000 to 2017-10-01T07:49:53.320.905.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.147016

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.0141277 to 23.0144167
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0933750  -- OK for SL --
   MIU time from 2017-10-01T07:50:09.467.921.000 to 2017-10-01T07:50:09.468.210.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.093426

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.1077917 to 37.1080807
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-01T07:50:23.561.636.000 to 2017-10-01T07:50:23.561.925.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-01T07:49:53.320.475.750
   Max SL corrected packet time: 2017-10-01T07:50:23.561.784.750
   Min SL corrected XO rel time: 5.3206173
   Max SL corrected XO rel time: 35.5619259
   TM values from -27 to 15 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-01T07:49:53.320.475.750 (XO: 5.3206173, t=34.0°C)  Rank: 00004
   TM values: Max =    11  Min =   -21  Ave =    -4.56  Std =     4.88
   Min TM value:-21 at an offset of 134.088 usec -> 2017-10-01T07:49:53.320.609.837 (XO: 5.3207513)
   Max TM value:11 at an offset of 119.625 usec -> 2017-10-01T07:49:53.320.595.375 (XO: 5.3207369)
   Max dTM/dt [1/us]::1.23e+02 at an offset of 118.400 usec -> 2017-10-01T07:49:53.320.594.150 (XO: 5.3207357)
   Min dTM/dt [1/us]::-1.21e+02 at an offset of 120.063 usec -> 2017-10-01T07:49:53.320.595.812 (XO: 5.3207373)

 - Block number: 2 starting at 2017-10-01T07:50:09.467.780.750 (XO: 21.4679227, t=34.0°C)  Rank: 00006
   TM values: Max =    15  Min =   -27  Ave =    -4.46  Std =     5.86
   Min TM value:-27 at an offset of 107.363 usec -> 2017-10-01T07:50:09.467.888.112 (XO: 21.4680301)
   Max TM value:15 at an offset of 106.375 usec -> 2017-10-01T07:50:09.467.887.125 (XO: 21.4680291)
   Max dTM/dt [1/us]::1.58e+02 at an offset of 120.988 usec -> 2017-10-01T07:50:09.467.901.737 (XO: 21.4680437)
   Min dTM/dt [1/us]::-1.61e+02 at an offset of 120.513 usec -> 2017-10-01T07:50:09.467.901.262 (XO: 21.4680432)

 - Block number: 3 starting at 2017-10-01T07:50:23.561.495.750 (XO: 35.5616369, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -20  Ave =    -4.50  Std =     4.64
   Min TM value:-20 at an offset of 143.463 usec -> 2017-10-01T07:50:23.561.639.213 (XO: 35.5617804)
   Max TM value:12 at an offset of 129.000 usec -> 2017-10-01T07:50:23.561.624.750 (XO: 35.5617659)
   Max dTM/dt [1/us]::1.39e+02 at an offset of 305.964 usec -> 2017-10-01T07:50:23.561.801.713 (XO: 35.5619429)
   Min dTM/dt [1/us]::-1.17e+02 at an offset of 104.288 usec -> 2017-10-01T07:50:23.561.600.037 (XO: 35.5617412)
