-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170928_092046_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.176197 to 44.603298
-each separately from 12s to 44s and from 129731us to 603298us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.953245 in 1 packets - #/No: 18/18
      16.472989 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-28T09:21:01.176.197.000 to 2017-09-28T09:21:33.603.298.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.953245 in 1 packets - #/No: 18/18
      16.472989 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4392041 to 45.4390816
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999387 in 1 packets - #/No: 18/18
      15.9999388 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170928_092046_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.6154021 to 13.6156911
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9531830  -- OK for SL --
   MIU time from 2017-09-28T09:21:01.176.197.000 to 2017-09-28T09:21:01.176.486.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.953245

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.5688741 to 29.5691631
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4729270  -- OK for SL --
   MIU time from 2017-09-28T09:21:17.129.731.000 to 2017-09-28T09:21:17.130.020.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.472989

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.0420901 to 46.0423791
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-28T09:21:33.603.009.000 to 2017-09-28T09:21:33.603.298.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-28T09:21:01.176.056.750
   Max SL corrected packet time: 2017-09-28T09:21:33.603.157.750
   Min SL corrected XO rel time: 12.1761987
   Max SL corrected XO rel time: 44.6033031
   TM values from -12 to 3 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-28T09:21:01.176.056.750 (XO: 12.1761987, t=34.0°C)  Rank: 00003
   TM values: Max =     3  Min =   -12  Ave =    -4.36  Std =     1.94
   Min TM value:-12 at an offset of 128.213 usec -> 2017-09-28T09:21:01.176.184.963 (XO: 12.1763269)
   Max TM value:3 at an offset of 141.101 usec -> 2017-09-28T09:21:01.176.197.850 (XO: 12.1763398)
   Max dTM/dt [1/us]::4.70e+01 at an offset of 191.263 usec -> 2017-09-28T09:21:01.176.248.013 (XO: 12.1763900)
   Min dTM/dt [1/us]::-4.70e+01 at an offset of 108.175 usec -> 2017-09-28T09:21:01.176.164.925 (XO: 12.1763069)

 - Block number: 2 starting at 2017-09-28T09:21:17.129.590.750 (XO: 28.1297318, t=34.0°C)  Rank: 00002
   TM values: Max =     1  Min =    -9  Ave =    -4.40  Std =     1.76
   Min TM value:-9 at an offset of 4.500 usec -> 2017-09-28T09:21:17.129.595.250 (XO: 28.1297363)
   Max TM value:1 at an offset of 104.288 usec -> 2017-09-28T09:21:17.129.695.037 (XO: 28.1298361)
   Max dTM/dt [1/us]::3.70e+01 at an offset of 62.525 usec -> 2017-09-28T09:21:17.129.653.275 (XO: 28.1297943)
   Min dTM/dt [1/us]::-3.80e+01 at an offset of 290.176 usec -> 2017-09-28T09:21:17.129.880.926 (XO: 28.1300220)

 - Block number: 3 starting at 2017-09-28T09:21:33.602.868.750 (XO: 44.6030141, t=34.0°C)  Rank: 00003
   TM values: Max =     3  Min =   -11  Ave =    -4.31  Std =     2.05
   Min TM value:-11 at an offset of 78.600 usec -> 2017-09-28T09:21:33.602.947.350 (XO: 44.6030927)
   Max TM value:3 at an offset of 67.413 usec -> 2017-09-28T09:21:33.602.936.162 (XO: 44.6030815)
   Max dTM/dt [1/us]::4.50e+01 at an offset of 270.626 usec -> 2017-09-28T09:21:33.603.139.376 (XO: 44.6032847)
   Min dTM/dt [1/us]::-4.10e+01 at an offset of 222.526 usec -> 2017-09-28T09:21:33.603.091.275 (XO: 44.6032366)
