-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170925_161357_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.065461 to 33.655234
-each separately from 1s to 33s and from 65461us to 655234us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.314083 in 1 packets - #/No: 18/18
      6.274823 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-25T16:14:00.065.461.000 to 2017-09-25T16:14:32.655.234.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.314083 in 1 packets - #/No: 18/18
      6.274823 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4373200 to 34.4371974
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      25.9999003 in 1 packets - #/No: 18/18
      5.9999771 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170925_161357_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.5027815 to 2.5030705
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.3139830  -- OK for SL --
   MIU time from 2017-09-25T16:14:00.065.461.000 to 2017-09-25T16:14:00.065.750.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.314083

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.8170535 to 28.8173425
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.2747990  -- OK for SL --
   MIU time from 2017-09-25T16:14:26.379.833.000 to 2017-09-25T16:14:26.380.122.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.274823

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.0921415 to 35.0924305
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-25T16:14:32.654.945.000 to 2017-09-25T16:14:32.655.234.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-25T16:14:00.065.320.750
   Max SL corrected packet time: 2017-09-25T16:14:32.655.093.750
   Min SL corrected XO rel time: 1.0654617
   Max SL corrected XO rel time: 33.6552357
   TM values from -30 to 25 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-25T16:14:00.065.320.750 (XO: 1.0654617, t=35.6°C)  Rank: 00003
   TM values: Max =    15  Min =   -23  Ave =    -4.51  Std =     5.46
   Min TM value:-23 at an offset of 100.463 usec -> 2017-09-25T16:14:00.065.421.212 (XO: 1.0655622)
   Max TM value:15 at an offset of 221.513 usec -> 2017-09-25T16:14:00.065.542.263 (XO: 1.0656833)
   Max dTM/dt [1/us]::1.90e+02 at an offset of 96.300 usec -> 2017-09-25T16:14:00.065.417.050 (XO: 1.0655580)
   Min dTM/dt [1/us]::-1.99e+02 at an offset of 8.675 usec -> 2017-09-25T16:14:00.065.329.425 (XO: 1.0654704)

 - Block number: 2 starting at 2017-09-25T16:14:26.379.692.750 (XO: 27.3798346, t=35.6°C)  Rank: 00006
   TM values: Max =    25  Min =   -30  Ave =    -4.59  Std =     7.64
   Min TM value:-30 at an offset of 168.851 usec -> 2017-09-25T16:14:26.379.861.600 (XO: 27.3800035)
   Max TM value:25 at an offset of 168.301 usec -> 2017-09-25T16:14:26.379.861.050 (XO: 27.3800029)
   Max dTM/dt [1/us]::2.39e+02 at an offset of 225.338 usec -> 2017-09-25T16:14:26.379.918.088 (XO: 27.3800600)
   Min dTM/dt [1/us]::-2.49e+02 at an offset of 150.213 usec -> 2017-09-25T16:14:26.379.842.963 (XO: 27.3799849)

 - Block number: 3 starting at 2017-09-25T16:14:32.654.804.750 (XO: 33.6549467, t=35.6°C)  Rank: 00006
   TM values: Max =    20  Min =   -29  Ave =    -4.52  Std =     7.32
   Min TM value:-29 at an offset of 128.551 usec -> 2017-09-25T16:14:32.654.933.300 (XO: 33.6550752)
   Max TM value:20 at an offset of 108.038 usec -> 2017-09-25T16:14:32.654.912.787 (XO: 33.6550547)
   Max dTM/dt [1/us]::2.53e+02 at an offset of 121.400 usec -> 2017-09-25T16:14:32.654.926.150 (XO: 33.6550681)
   Min dTM/dt [1/us]::-2.30e+02 at an offset of 70.388 usec -> 2017-09-25T16:14:32.654.875.137 (XO: 33.6550170)
