-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170925_144311_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.114926 to 30.617246
-each separately from 14s to 30s and from 114926us to 711904us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.596400 in 1 packets - #/No: 18/18
      11.905053 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-25T14:43:27.114.926.000 to 2017-09-25T14:43:43.617.246.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.596400 in 1 packets - #/No: 18/18
      11.905053 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4337032 to 31.4336420
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      3.9999847 in 1 packets - #/No: 18/18
      11.9999541 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170925_144311_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.5486296 to 15.5489186
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.5963830  -- OK for SL --
   MIU time from 2017-09-25T14:43:27.114.926.000 to 2017-09-25T14:43:27.115.215.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.596400

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.1453016 to 20.1455906
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9050070  -- OK for SL --
   MIU time from 2017-09-25T14:43:31.711.615.000 to 2017-09-25T14:43:31.711.904.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.905053

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.0505976 to 32.0508866
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-25T14:43:43.616.957.000 to 2017-09-25T14:43:43.617.246.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-25T14:43:27.114.785.750
   Max SL corrected packet time: 2017-09-25T14:43:43.617.105.750
   Min SL corrected XO rel time: 14.1149268
   Max SL corrected XO rel time: 30.6172470
   TM values from -20 to 10 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-25T14:43:27.114.785.750 (XO: 14.1149268, t=34.0°C)  Rank: 00003
   TM values: Max =     7  Min =   -15  Ave =    -4.44  Std =     2.75
   Min TM value:-15 at an offset of 186.788 usec -> 2017-09-25T14:43:27.114.972.538 (XO: 14.1151136)
   Max TM value:7 at an offset of 249.963 usec -> 2017-09-25T14:43:27.115.035.713 (XO: 14.1151768)
   Max dTM/dt [1/us]::8.20e+01 at an offset of 139.451 usec -> 2017-09-25T14:43:27.114.925.200 (XO: 14.1150663)
   Min dTM/dt [1/us]::-8.10e+01 at an offset of 233.688 usec -> 2017-09-25T14:43:27.115.019.438 (XO: 14.1151605)

 - Block number: 2 starting at 2017-09-25T14:43:31.711.474.750 (XO: 18.7116164, t=34.0°C)  Rank: 00002
   TM values: Max =    10  Min =   -19  Ave =    -4.49  Std =     4.15
   Min TM value:-19 at an offset of 290.264 usec -> 2017-09-25T14:43:31.711.765.013 (XO: 18.7119067)
   Max TM value:10 at an offset of 123.663 usec -> 2017-09-25T14:43:31.711.598.412 (XO: 18.7117401)
   Max dTM/dt [1/us]::1.09e+02 at an offset of 59.350 usec -> 2017-09-25T14:43:31.711.534.100 (XO: 18.7116758)
   Min dTM/dt [1/us]::-1.13e+02 at an offset of 7.125 usec -> 2017-09-25T14:43:31.711.481.875 (XO: 18.7116235)

 - Block number: 3 starting at 2017-09-25T14:43:43.616.816.750 (XO: 30.6169580, t=34.0°C)  Rank: 00002
   TM values: Max =    10  Min =   -20  Ave =    -4.42  Std =     4.17
   Min TM value:-20 at an offset of 169.301 usec -> 2017-09-25T14:43:43.616.986.050 (XO: 30.6171273)
   Max TM value:10 at an offset of 127.588 usec -> 2017-09-25T14:43:43.616.944.337 (XO: 30.6170856)
   Max dTM/dt [1/us]::9.80e+01 at an offset of 160.701 usec -> 2017-09-25T14:43:43.616.977.450 (XO: 30.6171187)
   Min dTM/dt [1/us]::-9.70e+01 at an offset of 117.125 usec -> 2017-09-25T14:43:43.616.933.875 (XO: 30.6170751)
