-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170916_061107_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.000087 to 44.759824
-each separately from 0s to 44s and from 87us to 759824us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.511977 in 1 packets - #/No: 18/18
      29.246893 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-16T06:11:10.000.087.000 to 2017-09-16T06:11:54.759.824.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.511977 in 1 packets - #/No: 18/18
      29.246893 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4335037 to 45.4333377
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999433 in 1 packets - #/No: 18/18
      28.9998907 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170916_061107_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4335916 to 1.4338806
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5119190  -- OK for SL --
   MIU time from 2017-09-16T06:11:10.000.087.000 to 2017-09-16T06:11:10.000.376.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.511977

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9457996 to 16.9460886
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.2467830  -- OK for SL --
   MIU time from 2017-09-16T06:11:25.512.353.000 to 2017-09-16T06:11:25.512.642.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.246893

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.1928716 to 46.1931606
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-16T06:11:54.759.535.000 to 2017-09-16T06:11:54.759.824.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-16T06:11:09.999.946.750
   Max SL corrected packet time: 2017-09-16T06:11:54.759.683.750
   Min SL corrected XO rel time: 0.0000879
   Max SL corrected XO rel time: 44.7598270
   TM values from -13 to 9 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-16T06:11:09.999.946.750 (XO: 0.0000879, t=34.0°C)  Rank: 00001
   TM values: Max =     2  Min =   -11  Ave =    -4.51  Std =     2.28
   Min TM value:-11 at an offset of 87.700 usec -> 2017-09-16T06:11:10.000.034.450 (XO: 0.0001756)
   Max TM value:2 at an offset of 21.813 usec -> 2017-09-16T06:11:09.999.968.562 (XO: 0.0001097)
   Max dTM/dt [1/us]::6.70e+01 at an offset of 75.888 usec -> 2017-09-16T06:11:10.000.022.637 (XO: 0.0001638)
   Min dTM/dt [1/us]::-7.00e+01 at an offset of 304.689 usec -> 2017-09-16T06:11:10.000.251.438 (XO: 0.0003926)

 - Block number: 2 starting at 2017-09-16T06:11:25.512.212.750 (XO: 15.5123545, t=34.0°C)  Rank: 00002
   TM values: Max =     4  Min =   -13  Ave =    -4.45  Std =     3.00
   Min TM value:-13 at an offset of 50.875 usec -> 2017-09-16T06:11:25.512.263.625 (XO: 15.5124054)
   Max TM value:4 at an offset of 49.988 usec -> 2017-09-16T06:11:25.512.262.737 (XO: 15.5124045)
   Max dTM/dt [1/us]::1.06e+02 at an offset of 84.675 usec -> 2017-09-16T06:11:25.512.297.425 (XO: 15.5124392)
   Min dTM/dt [1/us]::-1.01e+02 at an offset of 45.663 usec -> 2017-09-16T06:11:25.512.258.412 (XO: 15.5124002)

 - Block number: 3 starting at 2017-09-16T06:11:54.759.394.750 (XO: 44.7595380, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -13  Ave =    -4.40  Std =     2.23
   Min TM value:-13 at an offset of 285.276 usec -> 2017-09-16T06:11:54.759.680.026 (XO: 44.7598233)
   Max TM value:9 at an offset of 47.688 usec -> 2017-09-16T06:11:54.759.442.437 (XO: 44.7595857)
   Max dTM/dt [1/us]::8.10e+01 at an offset of 48.900 usec -> 2017-09-16T06:11:54.759.443.650 (XO: 44.7595869)
   Min dTM/dt [1/us]::-1.16e+02 at an offset of 47.738 usec -> 2017-09-16T06:11:54.759.442.487 (XO: 44.7595857)
