-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170914_001205_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.000096 to 35.064276
-each separately from 0s to 35s and from 96us to 387789us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.387115 in 1 packets - #/No: 18/18
      14.676198 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-14T00:12:08.000.096.000 to 2017-09-14T00:12:43.064.276.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.387115 in 1 packets - #/No: 18/18
      14.676198 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4355452 to 36.4354143
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.9999252 in 1 packets - #/No: 18/18
      14.9999439 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170914_001205_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4356427 to 1.4359317
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.3870390  -- OK for SL --
   MIU time from 2017-09-14T00:12:08.000.096.000 to 2017-09-14T00:12:08.000.385.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.387115

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.8229707 to 21.8232597
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6761430  -- OK for SL --
   MIU time from 2017-09-14T00:12:28.387.500.000 to 2017-09-14T00:12:28.387.789.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.676198

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.4994027 to 36.4996917
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-14T00:12:43.063.987.000 to 2017-09-14T00:12:43.064.276.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-14T00:12:07.999.955.750
   Max SL corrected packet time: 2017-09-14T00:12:43.064.135.750
   Min SL corrected XO rel time: 0.0000975
   Max SL corrected XO rel time: 35.0642776
   TM values from -12 to 8 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-14T00:12:07.999.955.750 (XO: 0.0000975, t=34.0°C)  Rank: 00001
   TM values: Max =     2  Min =   -12  Ave =    -4.54  Std =     1.95
   Min TM value:-12 at an offset of 208.576 usec -> 2017-09-14T00:12:08.000.164.325 (XO: 0.0003061)
   Max TM value:2 at an offset of 41.013 usec -> 2017-09-14T00:12:07.999.996.762 (XO: 0.0001385)
   Max dTM/dt [1/us]::7.20e+01 at an offset of 81.263 usec -> 2017-09-14T00:12:08.000.037.012 (XO: 0.0001788)
   Min dTM/dt [1/us]::-7.60e+01 at an offset of 217.376 usec -> 2017-09-14T00:12:08.000.173.125 (XO: 0.0003149)

 - Block number: 2 starting at 2017-09-14T00:12:28.387.359.750 (XO: 20.3875018, t=34.0°C)  Rank: 00003
   TM values: Max =     8  Min =   -12  Ave =    -4.57  Std =     2.08
   Min TM value:-12 at an offset of 50.038 usec -> 2017-09-14T00:12:28.387.409.787 (XO: 20.3875518)
   Max TM value:8 at an offset of 108.338 usec -> 2017-09-14T00:12:28.387.468.087 (XO: 20.3876101)
   Max dTM/dt [1/us]::8.50e+01 at an offset of 108.850 usec -> 2017-09-14T00:12:28.387.468.600 (XO: 20.3876106)
   Min dTM/dt [1/us]::-7.20e+01 at an offset of 108.375 usec -> 2017-09-14T00:12:28.387.468.125 (XO: 20.3876101)

 - Block number: 3 starting at 2017-09-14T00:12:43.063.846.750 (XO: 35.0639886, t=34.0°C)  Rank: 00002
   TM values: Max =     2  Min =   -11  Ave =    -4.62  Std =     1.89
   Min TM value:-11 at an offset of 138.588 usec -> 2017-09-14T00:12:43.063.985.338 (XO: 35.0641272)
   Max TM value:2 at an offset of 97.500 usec -> 2017-09-14T00:12:43.063.944.250 (XO: 35.0640861)
   Max dTM/dt [1/us]::5.70e+01 at an offset of 86.575 usec -> 2017-09-14T00:12:43.063.933.325 (XO: 35.0640752)
   Min dTM/dt [1/us]::-5.80e+01 at an offset of 173.876 usec -> 2017-09-14T00:12:43.064.020.625 (XO: 35.0641625)
