-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170913_173837_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.985363 to 31.303360
-each separately from 14s to 31s and from 29835us to 985652us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.044183 in 1 packets - #/No: 18/18
      14.272947 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-13T17:38:54.985.363.000 to 2017-09-13T17:39:11.303.360.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.044183 in 1 packets - #/No: 18/18
      14.272947 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4386091 to 32.4385469
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.9999890 in 1 packets - #/No: 18/18
      13.9999488 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170913_173837_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      19 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4239705 to 16.4242595
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.0441750  -- OK for SL --
   MIU time from 2017-09-13T17:38:54.985.363.000 to 2017-09-13T17:38:54.985.652.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.044183

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      23 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4684345 to 18.4687235
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2728950  -- OK for SL --
   MIU time from 2017-09-13T17:38:57.029.835.000 to 2017-09-13T17:38:57.030.124.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.272947

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.7416185 to 32.7419075
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-13T17:39:11.303.071.000 to 2017-09-13T17:39:11.303.360.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-13T17:38:54.985.222.750
   Max SL corrected packet time: 2017-09-13T17:39:11.303.219.750
   Min SL corrected XO rel time: 14.9853649
   Max SL corrected XO rel time: 31.3033617
   TM values from -64 to 54 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-13T17:38:54.985.222.750 (XO: 14.9853649, t=34.0°C)  Rank: 00019
   TM values: Max =    50  Min =   -59  Ave =    -4.64  Std =    18.79
   Min TM value:-59 at an offset of 124.175 usec -> 2017-09-13T17:38:54.985.346.925 (XO: 14.9854891)
   Max TM value:50 at an offset of 257.413 usec -> 2017-09-13T17:38:54.985.480.163 (XO: 14.9856224)
   Max dTM/dt [1/us]::5.12e+02 at an offset of 115.138 usec -> 2017-09-13T17:38:54.985.337.887 (XO: 14.9854801)
   Min dTM/dt [1/us]::-4.98e+02 at an offset of 124.125 usec -> 2017-09-13T17:38:54.985.346.875 (XO: 14.9854891)

 - Block number: 2 starting at 2017-09-13T17:38:57.029.694.750 (XO: 17.0298365, t=34.0°C)  Rank: 00023
   TM values: Max =    54  Min =   -64  Ave =    -4.64  Std =    19.29
   Min TM value:-64 at an offset of 159.651 usec -> 2017-09-13T17:38:57.029.854.400 (XO: 17.0299962)
   Max TM value:54 at an offset of 176.963 usec -> 2017-09-13T17:38:57.029.871.713 (XO: 17.0300135)
   Max dTM/dt [1/us]::5.40e+02 at an offset of 151.638 usec -> 2017-09-13T17:38:57.029.846.388 (XO: 17.0299881)
   Min dTM/dt [1/us]::-6.05e+02 at an offset of 160.438 usec -> 2017-09-13T17:38:57.029.855.188 (XO: 17.0299969)

 - Block number: 3 starting at 2017-09-13T17:39:11.302.930.750 (XO: 31.3030727, t=34.0°C)  Rank: 00008
   TM values: Max =    27  Min =   -39  Ave =    -4.52  Std =     9.51
   Min TM value:-39 at an offset of 127.300 usec -> 2017-09-13T17:39:11.303.058.050 (XO: 31.3032000)
   Max TM value:27 at an offset of 126.563 usec -> 2017-09-13T17:39:11.303.057.312 (XO: 31.3031993)
   Max dTM/dt [1/us]::3.03e+02 at an offset of 167.713 usec -> 2017-09-13T17:39:11.303.098.463 (XO: 31.3032404)
   Min dTM/dt [1/us]::-3.40e+02 at an offset of 125.900 usec -> 2017-09-13T17:39:11.303.056.650 (XO: 31.3031986)
