-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170912_085138_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.019599 to 31.780482
-each separately from 6s to 31s and from 19599us to 811744us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.791567 in 1 packets - #/No: 18/18
      8.968449 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-12T08:51:47.019.599.000 to 2017-09-12T08:52:12.780.482.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.791567 in 1 packets - #/No: 18/18
      8.968449 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4354876 to 32.4353928
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999394 in 1 packets - #/No: 18/18
      8.9999658 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170912_085138_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.4550882 to 7.4553772
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7915030  -- OK for SL --
   MIU time from 2017-09-12T08:51:47.019.599.000 to 2017-09-12T08:51:47.019.888.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.791567

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2468802 to 24.2471692
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9684150  -- OK for SL --
   MIU time from 2017-09-12T08:52:03.811.455.000 to 2017-09-12T08:52:03.811.744.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.968449

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.2155842 to 33.2158732
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-12T08:52:12.780.193.000 to 2017-09-12T08:52:12.780.482.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-12T08:51:47.019.458.750
   Max SL corrected packet time: 2017-09-12T08:52:12.780.341.750
   Min SL corrected XO rel time: 6.0196007
   Max SL corrected XO rel time: 31.7804834
   TM values from -30 to 21 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-12T08:51:47.019.458.750 (XO: 6.0196007, t=34.0°C)  Rank: 00005
   TM values: Max =    19  Min =   -29  Ave =    -4.70  Std =     7.05
   Min TM value:-29 at an offset of 156.288 usec -> 2017-09-12T08:51:47.019.615.038 (XO: 6.0197570)
   Max TM value:19 at an offset of 164.601 usec -> 2017-09-12T08:51:47.019.623.350 (XO: 6.0197653)
   Max dTM/dt [1/us]::1.49e+02 at an offset of 153.513 usec -> 2017-09-12T08:51:47.019.612.263 (XO: 6.0197542)
   Min dTM/dt [1/us]::-1.45e+02 at an offset of 157.551 usec -> 2017-09-12T08:51:47.019.616.300 (XO: 6.0197582)

 - Block number: 2 starting at 2017-09-12T08:52:03.811.314.750 (XO: 22.8114563, t=34.0°C)  Rank: 00005
   TM values: Max =    21  Min =   -27  Ave =    -4.58  Std =     7.31
   Min TM value:-27 at an offset of 191.026 usec -> 2017-09-12T08:52:03.811.505.775 (XO: 22.8116473)
   Max TM value:21 at an offset of 165.838 usec -> 2017-09-12T08:52:03.811.480.588 (XO: 22.8116221)
   Max dTM/dt [1/us]::1.55e+02 at an offset of 118.500 usec -> 2017-09-12T08:52:03.811.433.250 (XO: 22.8115748)
   Min dTM/dt [1/us]::-1.57e+02 at an offset of 120.500 usec -> 2017-09-12T08:52:03.811.435.250 (XO: 22.8115768)

 - Block number: 3 starting at 2017-09-12T08:52:12.780.052.750 (XO: 31.7801944, t=34.0°C)  Rank: 00004
   TM values: Max =    19  Min =   -30  Ave =    -4.57  Std =     6.74
   Min TM value:-30 at an offset of 169.726 usec -> 2017-09-12T08:52:12.780.222.475 (XO: 31.7803642)
   Max TM value:19 at an offset of 157.013 usec -> 2017-09-12T08:52:12.780.209.763 (XO: 31.7803515)
   Max dTM/dt [1/us]::1.65e+02 at an offset of 0.000 usec -> 2017-09-12T08:52:12.780.052.750 (XO: 31.7801944)
   Min dTM/dt [1/us]::-1.50e+02 at an offset of 91.713 usec -> 2017-09-12T08:52:12.780.144.462 (XO: 31.7802862)
