-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170909_120258_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.215422 to 32.684332
-each separately from 8s to 32s and from 215422us to 684332us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.232893 in 1 packets - #/No: 18/18
      12.235150 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-09T12:03:09.215.422.000 to 2017-09-09T12:03:33.684.332.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.232893 in 1 packets - #/No: 18/18
      12.235150 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4347775 to 33.4346869
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999548 in 1 packets - #/No: 18/18
      11.9999546 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170909_120258_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.6502004 to 9.6504894
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2328470  -- OK for SL --
   MIU time from 2017-09-09T12:03:09.215.422.000 to 2017-09-09T12:03:09.215.711.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.232893

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.8833364 to 21.8836254
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2351030  -- OK for SL --
   MIU time from 2017-09-09T12:03:21.448.604.000 to 2017-09-09T12:03:21.448.893.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.235150

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.1187284 to 34.1190174
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-09T12:03:33.684.043.000 to 2017-09-09T12:03:33.684.332.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-09T12:03:09.215.281.750
   Max SL corrected packet time: 2017-09-09T12:03:33.684.191.750
   Min SL corrected XO rel time: 8.2154237
   Max SL corrected XO rel time: 32.6843330
   TM values from -89 to 51 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-09T12:03:09.215.281.750 (XO: 8.2154237, t=34.0°C)  Rank: 00008
   TM values: Max =    51  Min =   -78  Ave =    -4.58  Std =     5.74
   Min TM value:-78 at an offset of 101.038 usec -> 2017-09-09T12:03:09.215.382.787 (XO: 8.2155247)
   Max TM value:51 at an offset of 100.738 usec -> 2017-09-09T12:03:09.215.382.487 (XO: 8.2155244)
   Max dTM/dt [1/us]::4.35e+02 at an offset of 100.675 usec -> 2017-09-09T12:03:09.215.382.425 (XO: 8.2155244)
   Min dTM/dt [1/us]::-6.02e+02 at an offset of 100.813 usec -> 2017-09-09T12:03:09.215.382.562 (XO: 8.2155245)

 - Block number: 2 starting at 2017-09-09T12:03:21.448.463.750 (XO: 20.4486058, t=34.0°C)  Rank: 00011
   TM values: Max =    40  Min =   -89  Ave =    -4.53  Std =     6.72
   Min TM value:-89 at an offset of 111.288 usec -> 2017-09-09T12:03:21.448.575.037 (XO: 20.4487170)
   Max TM value:40 at an offset of 113.188 usec -> 2017-09-09T12:03:21.448.576.937 (XO: 20.4487189)
   Max dTM/dt [1/us]::6.38e+02 at an offset of 110.900 usec -> 2017-09-09T12:03:21.448.574.650 (XO: 20.4487167)
   Min dTM/dt [1/us]::-5.97e+02 at an offset of 111.075 usec -> 2017-09-09T12:03:21.448.574.825 (XO: 20.4487168)

 - Block number: 3 starting at 2017-09-09T12:03:33.683.902.750 (XO: 32.6840440, t=34.0°C)  Rank: 00005
   TM values: Max =    28  Min =   -38  Ave =    -4.57  Std =     4.07
   Min TM value:-38 at an offset of 132.050 usec -> 2017-09-09T12:03:33.684.034.800 (XO: 32.6841761)
   Max TM value:28 at an offset of 131.825 usec -> 2017-09-09T12:03:33.684.034.575 (XO: 32.6841759)
   Max dTM/dt [1/us]::2.21e+02 at an offset of 132.100 usec -> 2017-09-09T12:03:33.684.034.850 (XO: 32.6841761)
   Min dTM/dt [1/us]::-3.36e+02 at an offset of 131.963 usec -> 2017-09-09T12:03:33.684.034.712 (XO: 32.6841760)
