-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170903_052756_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.823747 to 35.335668
-each separately from 14s to 35s and from 201940us to 824036us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.377904 in 1 packets - #/No: 18/18
      12.133150 in 1 packets - #/No: 36/18
MIU packet time from 2017-09-03T05:28:13.823.747.000 to 2017-09-03T05:28:34.335.668.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.377904 in 1 packets - #/No: 18/18
      12.133150 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4345261 to 36.4344442
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999649 in 1 packets - #/No: 18/18
      11.9999532 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170903_052756_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2582711 to 16.2585601
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3778710  -- OK for SL --
   MIU time from 2017-09-03T05:28:13.823.747.000 to 2017-09-03T05:28:13.824.036.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.377904

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.6364311 to 24.6367201
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1331030  -- OK for SL --
   MIU time from 2017-09-03T05:28:22.201.940.000 to 2017-09-03T05:28:22.202.229.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.133150

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.7698231 to 36.7701121
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-09-03T05:28:34.335.379.000 to 2017-09-03T05:28:34.335.668.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-09-03T05:28:13.823.606.750
   Max SL corrected packet time: 2017-09-03T05:28:34.335.527.750
   Min SL corrected XO rel time: 14.8237482
   Max SL corrected XO rel time: 35.3356692
   TM values from -42 to 32 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-09-03T05:28:13.823.606.750 (XO: 14.8237482, t=34.0°C)  Rank: 00006
   TM values: Max =    29  Min =   -37  Ave =    -4.62  Std =    10.27
   Min TM value:-37 at an offset of 109.325 usec -> 2017-09-03T05:28:13.823.716.075 (XO: 14.8238575)
   Max TM value:29 at an offset of 169.826 usec -> 2017-09-03T05:28:13.823.776.575 (XO: 14.8239180)
   Max dTM/dt [1/us]::4.60e+02 at an offset of 99.488 usec -> 2017-09-03T05:28:13.823.706.237 (XO: 14.8238477)
   Min dTM/dt [1/us]::-4.71e+02 at an offset of 109.288 usec -> 2017-09-03T05:28:13.823.716.037 (XO: 14.8238575)

 - Block number: 2 starting at 2017-09-03T05:28:22.201.799.750 (XO: 23.2019409, t=34.0°C)  Rank: 00008
   TM values: Max =    32  Min =   -42  Ave =    -4.48  Std =     9.06
   Min TM value:-42 at an offset of 99.463 usec -> 2017-09-03T05:28:22.201.899.212 (XO: 23.2020404)
   Max TM value:32 at an offset of 103.250 usec -> 2017-09-03T05:28:22.201.903.000 (XO: 23.2020441)
   Max dTM/dt [1/us]::4.07e+02 at an offset of 256.614 usec -> 2017-09-03T05:28:22.202.056.363 (XO: 23.2021975)
   Min dTM/dt [1/us]::-3.98e+02 at an offset of 96.163 usec -> 2017-09-03T05:28:22.201.895.912 (XO: 23.2020371)

 - Block number: 3 starting at 2017-09-03T05:28:34.335.238.750 (XO: 35.3353802, t=34.0°C)  Rank: 00008
   TM values: Max =    25  Min =   -34  Ave =    -4.49  Std =    11.67
   Min TM value:-34 at an offset of 118.563 usec -> 2017-09-03T05:28:34.335.357.312 (XO: 35.3354988)
   Max TM value:25 at an offset of 155.651 usec -> 2017-09-03T05:28:34.335.394.400 (XO: 35.3355359)
   Max dTM/dt [1/us]::4.84e+02 at an offset of 118.875 usec -> 2017-09-03T05:28:34.335.357.625 (XO: 35.3354991)
   Min dTM/dt [1/us]::-4.55e+02 at an offset of 118.125 usec -> 2017-09-03T05:28:34.335.356.875 (XO: 35.3354983)
