-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170826_040006_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.932739 to 30.042069
-each separately from 0s to 30s and from 41780us to 933028us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.226534 in 1 packets - #/No: 18/18
      14.881929 in 1 packets - #/No: 36/18
MIU packet time from 2017-08-26T04:00:09.932.739.000 to 2017-08-26T04:00:39.042.069.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.226534 in 1 packets - #/No: 18/18
      14.881929 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4348280 to 31.4347117
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999418 in 1 packets - #/No: 18/18
      14.9999419 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170826_040006_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3675646 to 2.3678536
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2264790  -- OK for SL --
   MIU time from 2017-08-26T04:00:09.932.739.000 to 2017-08-26T04:00:09.933.028.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.226534

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5943326 to 16.5946216
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8818710  -- OK for SL --
   MIU time from 2017-08-26T04:00:24.159.562.000 to 2017-08-26T04:00:24.159.851.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.881929

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4764926 to 31.4767816
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-08-26T04:00:39.041.780.000 to 2017-08-26T04:00:39.042.069.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-08-26T04:00:09.932.598.750
   Max SL corrected packet time: 2017-08-26T04:00:39.041.928.750
   Min SL corrected XO rel time: 0.9327402
   Max SL corrected XO rel time: 30.0420701
   TM values from -38 to 24 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-08-26T04:00:09.932.598.750 (XO: 0.9327402, t=34.0°C)  Rank: 00006
   TM values: Max =    24  Min =   -38  Ave =    -4.56  Std =     6.77
   Min TM value:-38 at an offset of 128.738 usec -> 2017-08-26T04:00:09.932.727.488 (XO: 0.9328690)
   Max TM value:24 at an offset of 132.401 usec -> 2017-08-26T04:00:09.932.731.150 (XO: 0.9328726)
   Max dTM/dt [1/us]::2.90e+02 at an offset of 132.288 usec -> 2017-08-26T04:00:09.932.731.038 (XO: 0.9328725)
   Min dTM/dt [1/us]::-2.47e+02 at an offset of 129.188 usec -> 2017-08-26T04:00:09.932.727.938 (XO: 0.9328694)

 - Block number: 2 starting at 2017-08-26T04:00:24.159.421.750 (XO: 15.1595634, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -21  Ave =    -4.51  Std =     4.88
   Min TM value:-21 at an offset of 124.313 usec -> 2017-08-26T04:00:24.159.546.062 (XO: 15.1596877)
   Max TM value:13 at an offset of 186.176 usec -> 2017-08-26T04:00:24.159.607.925 (XO: 15.1597496)
   Max dTM/dt [1/us]::1.39e+02 at an offset of 181.138 usec -> 2017-08-26T04:00:24.159.602.888 (XO: 15.1597446)
   Min dTM/dt [1/us]::-1.31e+02 at an offset of 297.426 usec -> 2017-08-26T04:00:24.159.719.176 (XO: 15.1598608)

 - Block number: 3 starting at 2017-08-26T04:00:39.041.639.750 (XO: 30.0417811, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -22  Ave =    -4.51  Std =     4.85
   Min TM value:-22 at an offset of 115.438 usec -> 2017-08-26T04:00:39.041.755.187 (XO: 30.0418965)
   Max TM value:13 at an offset of 134.351 usec -> 2017-08-26T04:00:39.041.774.100 (XO: 30.0419154)
   Max dTM/dt [1/us]::1.34e+02 at an offset of 135.413 usec -> 2017-08-26T04:00:39.041.775.163 (XO: 30.0419165)
   Min dTM/dt [1/us]::-1.35e+02 at an offset of 111.863 usec -> 2017-08-26T04:00:39.041.751.612 (XO: 30.0418929)
