-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170823_222709_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.964794 to 33.244346
-each separately from 8s to 33s and from 244057us to 965083us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.766081 in 1 packets - #/No: 18/18
      11.512604 in 1 packets - #/No: 36/18
MIU packet time from 2017-08-23T22:27:20.964.794.000 to 2017-08-23T22:27:45.244.346.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.766081 in 1 packets - #/No: 18/18
      11.512604 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4314166 to 34.4313193
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.9999494 in 1 packets - #/No: 18/18
      11.9999533 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170823_222709_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.3962087 to 10.3964977
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.7660310  -- OK for SL --
   MIU time from 2017-08-23T22:27:20.964.794.000 to 2017-08-23T22:27:20.965.083.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.766081

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.1625287 to 23.1628177
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.5125590  -- OK for SL --
   MIU time from 2017-08-23T22:27:33.731.164.000 to 2017-08-23T22:27:33.731.453.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.512604

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.6753767 to 34.6756657
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-08-23T22:27:45.244.057.000 to 2017-08-23T22:27:45.244.346.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-08-23T22:27:20.964.653.750
   Max SL corrected packet time: 2017-08-23T22:27:45.244.205.750
   Min SL corrected XO rel time: 8.9647959
   Max SL corrected XO rel time: 33.2443474
   TM values from -50 to 44 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-08-23T22:27:20.964.653.750 (XO: 8.9647959, t=34.0°C)  Rank: 00010
   TM values: Max =    38  Min =   -47  Ave =    -4.54  Std =    13.45
   Min TM value:-47 at an offset of 110.313 usec -> 2017-08-23T22:27:20.964.764.062 (XO: 8.9649062)
   Max TM value:38 at an offset of 131.613 usec -> 2017-08-23T22:27:20.964.785.363 (XO: 8.9649275)
   Max dTM/dt [1/us]::6.84e+02 at an offset of 117.813 usec -> 2017-08-23T22:27:20.964.771.562 (XO: 8.9649137)
   Min dTM/dt [1/us]::-6.71e+02 at an offset of 131.651 usec -> 2017-08-23T22:27:20.964.785.400 (XO: 8.9649275)

 - Block number: 2 starting at 2017-08-23T22:27:33.731.023.750 (XO: 21.7311656, t=34.0°C)  Rank: 00010
   TM values: Max =    42  Min =   -48  Ave =    -4.53  Std =    13.15
   Min TM value:-48 at an offset of 159.626 usec -> 2017-08-23T22:27:33.731.183.375 (XO: 21.7313252)
   Max TM value:42 at an offset of 160.501 usec -> 2017-08-23T22:27:33.731.184.250 (XO: 21.7313261)
   Max dTM/dt [1/us]::7.82e+02 at an offset of 180.876 usec -> 2017-08-23T22:27:33.731.204.625 (XO: 21.7313464)
   Min dTM/dt [1/us]::-7.48e+02 at an offset of 160.538 usec -> 2017-08-23T22:27:33.731.184.288 (XO: 21.7313261)

 - Block number: 3 starting at 2017-08-23T22:27:45.243.916.750 (XO: 33.2440584, t=34.0°C)  Rank: 00010
   TM values: Max =    44  Min =   -50  Ave =    -4.54  Std =    14.02
   Min TM value:-50 at an offset of 123.775 usec -> 2017-08-23T22:27:45.244.040.525 (XO: 33.2441821)
   Max TM value:44 at an offset of 134.563 usec -> 2017-08-23T22:27:45.244.051.313 (XO: 33.2441929)
   Max dTM/dt [1/us]::6.49e+02 at an offset of 107.300 usec -> 2017-08-23T22:27:45.244.024.050 (XO: 33.2441657)
   Min dTM/dt [1/us]::-6.36e+02 at an offset of 123.750 usec -> 2017-08-23T22:27:45.244.040.500 (XO: 33.2441821)
