-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170820_153949_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.437410 to 30.072401
-each separately from 1s to 30s and from 72112us to 437699us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.773640 in 1 packets - #/No: 18/18
      13.860484 in 1 packets - #/No: 36/18
MIU packet time from 2017-08-20T15:39:53.437.410.000 to 2017-08-20T15:40:22.072.401.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.773640 in 1 packets - #/No: 18/18
      13.860484 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4459907 to 31.4458796
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999425 in 1 packets - #/No: 18/18
      13.9999464 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170820_153949_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.8834004 to 2.8836894
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7735830  -- OK for SL --
   MIU time from 2017-08-20T15:39:53.437.410.000 to 2017-08-20T15:39:53.437.699.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.773640

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.6572724 to 17.6575614
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8604310  -- OK for SL --
   MIU time from 2017-08-20T15:40:08.211.339.000 to 2017-08-20T15:40:08.211.628.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.860484

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5179924 to 31.5182814
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-08-20T15:40:22.072.112.000 to 2017-08-20T15:40:22.072.401.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-08-20T15:39:53.437.269.750
   Max SL corrected packet time: 2017-08-20T15:40:22.072.260.750
   Min SL corrected XO rel time: 1.4374114
   Max SL corrected XO rel time: 30.0724021
   TM values from -30 to 22 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-08-20T15:39:53.437.269.750 (XO: 1.4374114, t=34.0°C)  Rank: 00004
   TM values: Max =    17  Min =   -26  Ave =    -4.47  Std =     5.89
   Min TM value:-26 at an offset of 79.963 usec -> 2017-08-20T15:39:53.437.349.712 (XO: 1.4374913)
   Max TM value:17 at an offset of 123.425 usec -> 2017-08-20T15:39:53.437.393.175 (XO: 1.4375348)
   Max dTM/dt [1/us]::1.75e+02 at an offset of 82.000 usec -> 2017-08-20T15:39:53.437.351.750 (XO: 1.4374934)
   Min dTM/dt [1/us]::-1.60e+02 at an offset of 75.225 usec -> 2017-08-20T15:39:53.437.344.975 (XO: 1.4374866)

 - Block number: 2 starting at 2017-08-20T15:40:08.211.198.750 (XO: 16.2113400, t=34.0°C)  Rank: 00004
   TM values: Max =    22  Min =   -30  Ave =    -4.40  Std =     7.13
   Min TM value:-30 at an offset of 129.863 usec -> 2017-08-20T15:40:08.211.328.612 (XO: 16.2114699)
   Max TM value:22 at an offset of 191.813 usec -> 2017-08-20T15:40:08.211.390.563 (XO: 16.2115318)
   Max dTM/dt [1/us]::1.79e+02 at an offset of 122.975 usec -> 2017-08-20T15:40:08.211.321.725 (XO: 16.2114630)
   Min dTM/dt [1/us]::-2.24e+02 at an offset of 178.526 usec -> 2017-08-20T15:40:08.211.377.275 (XO: 16.2115185)

 - Block number: 3 starting at 2017-08-20T15:40:22.071.971.750 (XO: 30.0721131, t=34.0°C)  Rank: 00003
   TM values: Max =    16  Min =   -23  Ave =    -4.35  Std =     5.34
   Min TM value:-23 at an offset of 270.201 usec -> 2017-08-20T15:40:22.072.241.951 (XO: 30.0723833)
   Max TM value:16 at an offset of 66.938 usec -> 2017-08-20T15:40:22.072.038.687 (XO: 30.0721800)
   Max dTM/dt [1/us]::1.25e+02 at an offset of 58.888 usec -> 2017-08-20T15:40:22.072.030.637 (XO: 30.0721720)
   Min dTM/dt [1/us]::-1.28e+02 at an offset of 67.638 usec -> 2017-08-20T15:40:22.072.039.387 (XO: 30.0721807)
