-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170820_063501_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.270544 to 33.257497
-each separately from 2s to 33s and from 257208us to 690893us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.419771 in 1 packets - #/No: 18/18
      15.566315 in 1 packets - #/No: 36/18
MIU packet time from 2017-08-20T06:35:06.270.544.000 to 2017-08-20T06:35:37.257.497.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.419771 in 1 packets - #/No: 18/18
      15.566315 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4346267 to 34.4345068
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999420 in 1 packets - #/No: 18/18
      15.9999381 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170820_063501_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7051711 to 3.7054601
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4197110  -- OK for SL --
   MIU time from 2017-08-20T06:35:06.270.544.000 to 2017-08-20T06:35:06.270.833.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.419771

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1251711 to 19.1254601
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5662550  -- OK for SL --
   MIU time from 2017-08-20T06:35:21.690.604.000 to 2017-08-20T06:35:21.690.893.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.566315

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.6917151 to 34.6920041
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-08-20T06:35:37.257.208.000 to 2017-08-20T06:35:37.257.497.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-08-20T06:35:06.270.403.750
   Max SL corrected packet time: 2017-08-20T06:35:37.257.356.750
   Min SL corrected XO rel time: 2.2705455
   Max SL corrected XO rel time: 33.2574983
   TM values from -67 to 49 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-08-20T06:35:06.270.403.750 (XO: 2.2705455, t=34.0°C)  Rank: 00015
   TM values: Max =    49  Min =   -67  Ave =    -4.61  Std =    15.99
   Min TM value:-67 at an offset of 156.851 usec -> 2017-08-20T06:35:06.270.560.600 (XO: 2.2707023)
   Max TM value:49 at an offset of 148.251 usec -> 2017-08-20T06:35:06.270.552.000 (XO: 2.2706937)
   Max dTM/dt [1/us]::3.07e+02 at an offset of 163.201 usec -> 2017-08-20T06:35:06.270.566.950 (XO: 2.2707087)
   Min dTM/dt [1/us]::-3.19e+02 at an offset of 127.588 usec -> 2017-08-20T06:35:06.270.531.338 (XO: 2.2706731)

 - Block number: 2 starting at 2017-08-20T06:35:21.690.463.750 (XO: 17.6906052, t=34.0°C)  Rank: 00015
   TM values: Max =    45  Min =   -51  Ave =    -4.58  Std =    16.18
   Min TM value:-51 at an offset of 284.514 usec -> 2017-08-20T06:35:21.690.748.263 (XO: 17.6908897)
   Max TM value:45 at an offset of 280.451 usec -> 2017-08-20T06:35:21.690.744.201 (XO: 17.6908856)
   Max dTM/dt [1/us]::2.52e+02 at an offset of 186.763 usec -> 2017-08-20T06:35:21.690.650.513 (XO: 17.6907919)
   Min dTM/dt [1/us]::-2.77e+02 at an offset of 178.238 usec -> 2017-08-20T06:35:21.690.641.988 (XO: 17.6907834)

 - Block number: 3 starting at 2017-08-20T06:35:37.257.067.750 (XO: 33.2572093, t=34.0°C)  Rank: 00011
   TM values: Max =    32  Min =   -40  Ave =    -4.50  Std =    10.47
   Min TM value:-40 at an offset of 162.901 usec -> 2017-08-20T06:35:37.257.230.650 (XO: 33.2573722)
   Max TM value:32 at an offset of 88.850 usec -> 2017-08-20T06:35:37.257.156.600 (XO: 33.2572982)
   Max dTM/dt [1/us]::3.40e+02 at an offset of 305.989 usec -> 2017-08-20T06:35:37.257.373.738 (XO: 33.2575153)
   Min dTM/dt [1/us]::-1.86e+02 at an offset of 141.101 usec -> 2017-08-20T06:35:37.257.208.850 (XO: 33.2573504)
