-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170809_134309_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.666852 to 37.807199
-each separately from 1s to 37s and from 282080us to 807199us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.614939 in 1 packets - #/No: 18/18
      20.524541 in 1 packets - #/No: 36/18
MIU packet time from 2017-08-09T13:43:13.666.852.000 to 2017-08-09T13:43:49.807.199.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.614939 in 1 packets - #/No: 18/18
      20.524541 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4324632 to 38.4323251
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999386 in 1 packets - #/No: 18/18
      19.9999233 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170809_134309_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.0993142 to 3.0996032
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6148790  -- OK for SL --
   MIU time from 2017-08-09T13:43:13.666.852.000 to 2017-08-09T13:43:13.667.141.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.614939

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.7144822 to 18.7147712
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5244630  -- OK for SL --
   MIU time from 2017-08-09T13:43:29.282.080.000 to 2017-08-09T13:43:29.282.369.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.524541

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.2392342 to 39.2395232
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-08-09T13:43:49.806.910.000 to 2017-08-09T13:43:49.807.199.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-08-09T13:43:13.666.711.750
   Max SL corrected packet time: 2017-08-09T13:43:49.807.058.750
   Min SL corrected XO rel time: 1.6668536
   Max SL corrected XO rel time: 37.8072012
   TM values from -72 to 40 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-08-09T13:43:13.666.711.750 (XO: 1.6668536, t=35.6°C)  Rank: 00006
   TM values: Max =    39  Min =   -48  Ave =    -4.48  Std =     5.26
   Min TM value:-48 at an offset of 90.963 usec -> 2017-08-09T13:43:13.666.802.712 (XO: 1.6669446)
   Max TM value:39 at an offset of 271.989 usec -> 2017-08-09T13:43:13.666.983.738 (XO: 1.6671256)
   Max dTM/dt [1/us]::3.12e+02 at an offset of 271.926 usec -> 2017-08-09T13:43:13.666.983.676 (XO: 1.6671255)
   Min dTM/dt [1/us]::-5.00e+02 at an offset of 272.039 usec -> 2017-08-09T13:43:13.666.983.788 (XO: 1.6671256)

 - Block number: 2 starting at 2017-08-09T13:43:29.281.939.750 (XO: 17.2820815, t=35.6°C)  Rank: 00010
   TM values: Max =    40  Min =   -54  Ave =    -4.39  Std =     6.81
   Min TM value:-54 at an offset of 47.100 usec -> 2017-08-09T13:43:29.281.986.850 (XO: 17.2821286)
   Max TM value:40 at an offset of 105.413 usec -> 2017-08-09T13:43:29.282.045.162 (XO: 17.2821869)
   Max dTM/dt [1/us]::3.99e+02 at an offset of 188.338 usec -> 2017-08-09T13:43:29.282.128.088 (XO: 17.2822698)
   Min dTM/dt [1/us]::-3.39e+02 at an offset of 183.526 usec -> 2017-08-09T13:43:29.282.123.275 (XO: 17.2822650)

 - Block number: 3 starting at 2017-08-09T13:43:49.806.769.750 (XO: 37.8069122, t=35.6°C)  Rank: 00012
   TM values: Max =    35  Min =   -72  Ave =    -4.61  Std =     8.72
   Min TM value:-72 at an offset of 84.825 usec -> 2017-08-09T13:43:49.806.854.575 (XO: 37.8069970)
   Max TM value:35 at an offset of 85.263 usec -> 2017-08-09T13:43:49.806.855.012 (XO: 37.8069974)
   Max dTM/dt [1/us]::4.83e+02 at an offset of 94.375 usec -> 2017-08-09T13:43:49.806.864.125 (XO: 37.8070065)
   Min dTM/dt [1/us]::-4.12e+02 at an offset of 150.038 usec -> 2017-08-09T13:43:49.806.919.788 (XO: 37.8070622)
