-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170730_075129_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.816577 to 30.078166
-each separately from 12s to 30s and from 77877us to 816866us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.821786 in 1 packets - #/No: 18/18
      14.438936 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-30T07:51:44.816.577.000 to 2017-07-30T07:52:02.078.166.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.821786 in 1 packets - #/No: 18/18
      14.438936 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4378799 to 31.4378103
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.9999884 in 1 packets - #/No: 18/18
      14.9999420 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170730_075129_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      24 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.2544557 to 14.2547447
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.8217750  -- OK for SL --
   MIU time from 2017-07-30T07:51:44.816.577.000 to 2017-07-30T07:51:44.816.866.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.821786

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      24 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0765197 to 17.0768087
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4388790  -- OK for SL --
   MIU time from 2017-07-30T07:51:47.638.652.000 to 2017-07-30T07:51:47.638.941.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.438936

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5156877 to 31.5159767
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-30T07:52:02.077.877.000 to 2017-07-30T07:52:02.078.166.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-30T07:51:44.816.436.750
   Max SL corrected packet time: 2017-07-30T07:52:02.078.025.750
   Min SL corrected XO rel time: 12.8165789
   Max SL corrected XO rel time: 30.0781667
   TM values from -56 to 47 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-30T07:51:44.816.436.750 (XO: 12.8165789, t=34.0°C)  Rank: 00024
   TM values: Max =    45  Min =   -53  Ave =    -4.43  Std =    19.66
   Min TM value:-53 at an offset of 128.713 usec -> 2017-07-30T07:51:44.816.565.462 (XO: 12.8167076)
   Max TM value:45 at an offset of 115.938 usec -> 2017-07-30T07:51:44.816.552.687 (XO: 12.8166948)
   Max dTM/dt [1/us]::3.77e+02 at an offset of 125.938 usec -> 2017-07-30T07:51:44.816.562.687 (XO: 12.8167048)
   Min dTM/dt [1/us]::-3.71e+02 at an offset of 133.001 usec -> 2017-07-30T07:51:44.816.569.750 (XO: 12.8167119)

 - Block number: 2 starting at 2017-07-30T07:51:47.638.511.750 (XO: 15.6386539, t=34.0°C)  Rank: 00024
   TM values: Max =    47  Min =   -56  Ave =    -4.50  Std =    20.03
   Min TM value:-56 at an offset of 159.001 usec -> 2017-07-30T07:51:47.638.670.750 (XO: 15.6388129)
   Max TM value:47 at an offset of 137.951 usec -> 2017-07-30T07:51:47.638.649.700 (XO: 15.6387918)
   Max dTM/dt [1/us]::3.68e+02 at an offset of 125.550 usec -> 2017-07-30T07:51:47.638.637.300 (XO: 15.6387794)
   Min dTM/dt [1/us]::-3.77e+02 at an offset of 128.013 usec -> 2017-07-30T07:51:47.638.639.762 (XO: 15.6387819)

 - Block number: 3 starting at 2017-07-30T07:52:02.077.736.750 (XO: 30.0778777, t=34.0°C)  Rank: 00010
   TM values: Max =    28  Min =   -38  Ave =    -4.46  Std =    10.84
   Min TM value:-38 at an offset of 163.188 usec -> 2017-07-30T07:52:02.077.899.938 (XO: 30.0780409)
   Max TM value:28 at an offset of 133.538 usec -> 2017-07-30T07:52:02.077.870.288 (XO: 30.0780112)
   Max dTM/dt [1/us]::2.11e+02 at an offset of 132.763 usec -> 2017-07-30T07:52:02.077.869.513 (XO: 30.0780105)
   Min dTM/dt [1/us]::-2.36e+02 at an offset of 305.989 usec -> 2017-07-30T07:52:02.078.042.738 (XO: 30.0781837)
