-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170727_031001_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.387164 to 33.481231
-each separately from 12s to 33s and from 387164us to 896757us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.509015 in 1 packets - #/No: 18/18
      6.584185 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-27T03:10:16.387.164.000 to 2017-07-27T03:10:37.481.231.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.509015 in 1 packets - #/No: 18/18
      6.584185 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4327393 to 34.4326581
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999459 in 1 packets - #/No: 18/18
      6.9999729 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170727_031001_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.8199033 to 13.8201923
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5089590  -- OK for SL --
   MIU time from 2017-07-27T03:10:16.387.164.000 to 2017-07-27T03:10:16.387.453.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.509015

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.3291513 to 28.3294403
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.5841590  -- OK for SL --
   MIU time from 2017-07-27T03:10:30.896.468.000 to 2017-07-27T03:10:30.896.757.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.584185

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.9135993 to 34.9138883
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-27T03:10:37.480.942.000 to 2017-07-27T03:10:37.481.231.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-27T03:10:16.387.023.750
   Max SL corrected packet time: 2017-07-27T03:10:37.481.090.750
   Min SL corrected XO rel time: 12.3871655
   Max SL corrected XO rel time: 33.4812321
   TM values from -35 to 25 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-27T03:10:16.387.023.750 (XO: 12.3871655, t=34.0°C)  Rank: 00011
   TM values: Max =    25  Min =   -34  Ave =    -4.39  Std =     9.18
   Min TM value:-34 at an offset of 159.251 usec -> 2017-07-27T03:10:16.387.183.000 (XO: 12.3873248)
   Max TM value:25 at an offset of 125.450 usec -> 2017-07-27T03:10:16.387.149.200 (XO: 12.3872910)
   Max dTM/dt [1/us]::5.06e+02 at an offset of 144.326 usec -> 2017-07-27T03:10:16.387.168.075 (XO: 12.3873098)
   Min dTM/dt [1/us]::-5.41e+02 at an offset of 159.213 usec -> 2017-07-27T03:10:16.387.182.963 (XO: 12.3873247)

 - Block number: 2 starting at 2017-07-27T03:10:30.896.327.750 (XO: 26.8964696, t=34.0°C)  Rank: 00009
   TM values: Max =    25  Min =   -35  Ave =    -4.40  Std =     8.71
   Min TM value:-35 at an offset of 135.401 usec -> 2017-07-27T03:10:30.896.463.150 (XO: 26.8966050)
   Max TM value:25 at an offset of 122.388 usec -> 2017-07-27T03:10:30.896.450.137 (XO: 26.8965920)
   Max dTM/dt [1/us]::5.18e+02 at an offset of 170.513 usec -> 2017-07-27T03:10:30.896.498.263 (XO: 26.8966401)
   Min dTM/dt [1/us]::-5.27e+02 at an offset of 122.425 usec -> 2017-07-27T03:10:30.896.450.175 (XO: 26.8965920)

 - Block number: 3 starting at 2017-07-27T03:10:37.480.801.750 (XO: 33.4809431, t=34.0°C)  Rank: 00011
   TM values: Max =    25  Min =   -33  Ave =    -4.46  Std =    10.42
   Min TM value:-33 at an offset of 166.413 usec -> 2017-07-27T03:10:37.480.968.163 (XO: 33.4811095)
   Max TM value:25 at an offset of 166.826 usec -> 2017-07-27T03:10:37.480.968.575 (XO: 33.4811099)
   Max dTM/dt [1/us]::4.94e+02 at an offset of 158.626 usec -> 2017-07-27T03:10:37.480.960.375 (XO: 33.4811017)
   Min dTM/dt [1/us]::-4.93e+02 at an offset of 158.538 usec -> 2017-07-27T03:10:37.480.960.288 (XO: 33.4811016)
