-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170726_163329_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.458790 to 34.401967
-each separately from 3s to 34s and from 401678us to 459079us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.986702 in 1 packets - #/No: 18/18
      18.955608 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-26T16:33:35.458.790.000 to 2017-07-26T16:34:06.401.967.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.986702 in 1 packets - #/No: 18/18
      18.955608 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4356910 to 35.4355708
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999534 in 1 packets - #/No: 18/18
      18.9999264 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170726_163329_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8944809 to 4.8947699
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9866550  -- OK for SL --
   MIU time from 2017-07-26T16:33:35.458.790.000 to 2017-07-26T16:33:35.459.079.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.986702

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8814249 to 16.8817139
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.9555350  -- OK for SL --
   MIU time from 2017-07-26T16:33:47.445.781.000 to 2017-07-26T16:33:47.446.070.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.955608

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.8372489 to 35.8375379
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-26T16:34:06.401.678.000 to 2017-07-26T16:34:06.401.967.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-26T16:33:35.458.649.750
   Max SL corrected packet time: 2017-07-26T16:34:06.401.826.750
   Min SL corrected XO rel time: 3.4587917
   Max SL corrected XO rel time: 34.4019687
   TM values from -41 to 33 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-26T16:33:35.458.649.750 (XO: 3.4587917, t=34.0°C)  Rank: 00008
   TM values: Max =    33  Min =   -41  Ave =    -4.52  Std =     9.33
   Min TM value:-41 at an offset of 121.513 usec -> 2017-07-26T16:33:35.458.771.262 (XO: 3.4589132)
   Max TM value:33 at an offset of 101.000 usec -> 2017-07-26T16:33:35.458.750.750 (XO: 3.4588927)
   Max dTM/dt [1/us]::3.36e+02 at an offset of 0.000 usec -> 2017-07-26T16:33:35.458.649.750 (XO: 3.4587917)
   Min dTM/dt [1/us]::-3.31e+02 at an offset of 121.138 usec -> 2017-07-26T16:33:35.458.770.887 (XO: 3.4589128)

 - Block number: 2 starting at 2017-07-26T16:33:47.445.640.750 (XO: 15.4457822, t=34.0°C)  Rank: 00006
   TM values: Max =    28  Min =   -37  Ave =    -4.48  Std =     8.51
   Min TM value:-37 at an offset of 117.288 usec -> 2017-07-26T16:33:47.445.758.037 (XO: 15.4458995)
   Max TM value:28 at an offset of 118.163 usec -> 2017-07-26T16:33:47.445.758.912 (XO: 15.4459004)
   Max dTM/dt [1/us]::3.48e+02 at an offset of 192.351 usec -> 2017-07-26T16:33:47.445.833.100 (XO: 15.4459745)
   Min dTM/dt [1/us]::-3.05e+02 at an offset of 191.463 usec -> 2017-07-26T16:33:47.445.832.213 (XO: 15.4459737)

 - Block number: 3 starting at 2017-07-26T16:34:06.401.537.750 (XO: 34.4016797, t=34.0°C)  Rank: 00008
   TM values: Max =    31  Min =   -38  Ave =    -4.53  Std =     9.90
   Min TM value:-38 at an offset of 142.388 usec -> 2017-07-26T16:34:06.401.680.138 (XO: 34.4018221)
   Max TM value:31 at an offset of 217.501 usec -> 2017-07-26T16:34:06.401.755.250 (XO: 34.4018972)
   Max dTM/dt [1/us]::3.04e+02 at an offset of 190.951 usec -> 2017-07-26T16:34:06.401.728.700 (XO: 34.4018707)
   Min dTM/dt [1/us]::-3.16e+02 at an offset of 205.638 usec -> 2017-07-26T16:34:06.401.743.388 (XO: 34.4018853)
