-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170724_090248_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 9.092186 to 39.485326
-each separately from 9s to 39s and from 92186us to 485326us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.246238 in 1 packets - #/No: 18/18
      18.146035 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-24T09:03:00.092.186.000 to 2017-07-24T09:03:30.485.326.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.246238 in 1 packets - #/No: 18/18
      18.146035 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4444689 to 40.4443549
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999544 in 1 packets - #/No: 18/18
      17.9999316 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170724_090248_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5366559 to 10.5369449
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2461910  -- OK for SL --
   MIU time from 2017-07-24T09:03:00.092.186.000 to 2017-07-24T09:03:00.092.475.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.246238

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7831359 to 22.7834249
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.1459670  -- OK for SL --
   MIU time from 2017-07-24T09:03:12.338.713.000 to 2017-07-24T09:03:12.339.002.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.146035

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      23 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.9293919 to 40.9296809
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-24T09:03:30.485.037.000 to 2017-07-24T09:03:30.485.326.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-24T09:03:00.092.045.750
   Max SL corrected packet time: 2017-07-24T09:03:30.485.185.750
   Min SL corrected XO rel time: 9.0921874
   Max SL corrected XO rel time: 39.4853278
   TM values from -83 to 82 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-24T09:03:00.092.045.750 (XO: 9.0921874, t=34.0°C)  Rank: 00009
   TM values: Max =    82  Min =   -83  Ave =    -4.43  Std =     4.75
   Min TM value:-83 at an offset of 105.113 usec -> 2017-07-24T09:03:00.092.150.862 (XO: 9.0922925)
   Max TM value:82 at an offset of 105.538 usec -> 2017-07-24T09:03:00.092.151.287 (XO: 9.0922929)
   Max dTM/dt [1/us]::5.06e+02 at an offset of 105.388 usec -> 2017-07-24T09:03:00.092.151.137 (XO: 9.0922927)
   Min dTM/dt [1/us]::-6.47e+02 at an offset of 105.063 usec -> 2017-07-24T09:03:00.092.150.812 (XO: 9.0922924)

 - Block number: 2 starting at 2017-07-24T09:03:12.338.572.750 (XO: 21.3387139, t=34.0°C)  Rank: 00015
   TM values: Max =    75  Min =   -41  Ave =    -4.39  Std =     5.97
   Min TM value:-41 at an offset of 106.788 usec -> 2017-07-24T09:03:12.338.679.537 (XO: 21.3388207)
   Max TM value:75 at an offset of 124.213 usec -> 2017-07-24T09:03:12.338.696.962 (XO: 21.3388381)
   Max dTM/dt [1/us]::4.81e+02 at an offset of 124.175 usec -> 2017-07-24T09:03:12.338.696.925 (XO: 21.3388381)
   Min dTM/dt [1/us]::-6.48e+02 at an offset of 106.675 usec -> 2017-07-24T09:03:12.338.679.425 (XO: 21.3388206)

 - Block number: 3 starting at 2017-07-24T09:03:30.484.896.750 (XO: 39.4850388, t=34.0°C)  Rank: 00023
   TM values: Max =    35  Min =   -44  Ave =    -4.39  Std =     6.80
   Min TM value:-44 at an offset of 97.213 usec -> 2017-07-24T09:03:30.484.993.962 (XO: 39.4851361)
   Max TM value:35 at an offset of 102.188 usec -> 2017-07-24T09:03:30.484.998.937 (XO: 39.4851410)
   Max dTM/dt [1/us]::2.11e+02 at an offset of 120.788 usec -> 2017-07-24T09:03:30.485.017.537 (XO: 39.4851596)
   Min dTM/dt [1/us]::-2.97e+02 at an offset of 121.550 usec -> 2017-07-24T09:03:30.485.018.300 (XO: 39.4851604)
