-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170721_110630_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.572240 to 30.328288
-each separately from 5s to 30s and from 312653us to 572529us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.740124 in 1 packets - #/No: 18/18
      9.015057 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-21T11:06:38.572.240.000 to 2017-07-21T11:07:03.328.288.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.740124 in 1 packets - #/No: 18/18
      9.015057 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4385841 to 31.4384884
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999389 in 1 packets - #/No: 18/18
      8.9999654 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170721_110630_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.0108238 to 7.0111128
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7400630  -- OK for SL --
   MIU time from 2017-07-21T11:06:38.572.240.000 to 2017-07-21T11:06:38.572.529.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.740124

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7511758 to 22.7514648
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.0150230  -- OK for SL --
   MIU time from 2017-07-21T11:06:54.312.653.000 to 2017-07-21T11:06:54.312.942.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.015057

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7664878 to 31.7667768
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-21T11:07:03.327.999.000 to 2017-07-21T11:07:03.328.288.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-21T11:06:38.572.099.750
   Max SL corrected packet time: 2017-07-21T11:07:03.328.147.750
   Min SL corrected XO rel time: 5.5722419
   Max SL corrected XO rel time: 30.3282896
   TM values from -26 to 20 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-21T11:06:38.572.099.750 (XO: 5.5722419, t=34.0°C)  Rank: 00004
   TM values: Max =    11  Min =   -21  Ave =    -4.67  Std =     4.58
   Min TM value:-21 at an offset of 147.713 usec -> 2017-07-21T11:06:38.572.247.463 (XO: 5.5723896)
   Max TM value:11 at an offset of 130.238 usec -> 2017-07-21T11:06:38.572.229.987 (XO: 5.5723721)
   Max dTM/dt [1/us]::1.08e+02 at an offset of 143.538 usec -> 2017-07-21T11:06:38.572.243.288 (XO: 5.5723854)
   Min dTM/dt [1/us]::-1.07e+02 at an offset of 110.600 usec -> 2017-07-21T11:06:38.572.210.350 (XO: 5.5723525)

 - Block number: 2 starting at 2017-07-21T11:06:54.312.512.750 (XO: 21.3126540, t=34.0°C)  Rank: 00008
   TM values: Max =    20  Min =   -26  Ave =    -4.57  Std =     6.16
   Min TM value:-26 at an offset of 119.025 usec -> 2017-07-21T11:06:54.312.631.775 (XO: 21.3127730)
   Max TM value:20 at an offset of 98.500 usec -> 2017-07-21T11:06:54.312.611.250 (XO: 21.3127525)
   Max dTM/dt [1/us]::1.29e+02 at an offset of 166.801 usec -> 2017-07-21T11:06:54.312.679.550 (XO: 21.3128208)
   Min dTM/dt [1/us]::-1.22e+02 at an offset of 166.238 usec -> 2017-07-21T11:06:54.312.678.988 (XO: 21.3128202)

 - Block number: 3 starting at 2017-07-21T11:07:03.327.858.750 (XO: 30.3280006, t=34.0°C)  Rank: 00005
   TM values: Max =    15  Min =   -24  Ave =    -4.47  Std =     6.66
   Min TM value:-24 at an offset of 177.363 usec -> 2017-07-21T11:07:03.328.036.113 (XO: 30.3281780)
   Max TM value:15 at an offset of 138.476 usec -> 2017-07-21T11:07:03.327.997.225 (XO: 30.3281391)
   Max dTM/dt [1/us]::1.17e+02 at an offset of 203.626 usec -> 2017-07-21T11:07:03.328.062.375 (XO: 30.3282043)
   Min dTM/dt [1/us]::-1.21e+02 at an offset of 166.476 usec -> 2017-07-21T11:07:03.328.025.225 (XO: 30.3281671)
