-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170719_125639_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.035128 to 39.987408
-each separately from 4s to 39s and from 35128us to 987408us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.140623 in 1 packets - #/No: 18/18
      18.810790 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-19T12:56:46.035.128.000 to 2017-07-19T12:57:21.987.408.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.140623 in 1 packets - #/No: 18/18
      18.810790 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.4360527 to 40.4359208
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999360 in 1 packets - #/No: 18/18
      17.9999321 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170719_125639_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.4711815 to 5.4714705
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1405590  -- OK for SL --
   MIU time from 2017-07-19T12:56:46.035.128.000 to 2017-07-19T12:56:46.035.417.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.140623

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.6120295 to 22.6123185
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8107190  -- OK for SL --
   MIU time from 2017-07-19T12:57:03.176.040.000 to 2017-07-19T12:57:03.176.329.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.810790

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.4230375 to 41.4233265
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-19T12:57:21.987.119.000 to 2017-07-19T12:57:21.987.408.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-19T12:56:46.034.987.750
   Max SL corrected packet time: 2017-07-19T12:57:21.987.267.750
   Min SL corrected XO rel time: 4.0351289
   Max SL corrected XO rel time: 39.9874096
   TM values from -126 to 78 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-19T12:56:46.034.987.750 (XO: 4.0351289, t=34.0°C)  Rank: 00004
   TM values: Max =    66  Min =   -32  Ave =    -4.41  Std =     3.72
   Min TM value:-32 at an offset of 126.275 usec -> 2017-07-19T12:56:46.035.114.025 (XO: 4.0352552)
   Max TM value:66 at an offset of 126.075 usec -> 2017-07-19T12:56:46.035.113.825 (XO: 4.0352550)
   Max dTM/dt [1/us]::5.47e+02 at an offset of 126.025 usec -> 2017-07-19T12:56:46.035.113.775 (XO: 4.0352550)
   Min dTM/dt [1/us]::-7.15e+02 at an offset of 126.125 usec -> 2017-07-19T12:56:46.035.113.875 (XO: 4.0352551)

 - Block number: 2 starting at 2017-07-19T12:57:03.175.899.750 (XO: 21.1760415, t=34.0°C)  Rank: 00007
   TM values: Max =    65  Min =   -55  Ave =    -4.48  Std =     5.68
   Min TM value:-55 at an offset of 111.113 usec -> 2017-07-19T12:57:03.176.010.862 (XO: 21.1761526)
   Max TM value:65 at an offset of 107.425 usec -> 2017-07-19T12:57:03.176.007.175 (XO: 21.1761489)
   Max dTM/dt [1/us]::5.61e+02 at an offset of 110.863 usec -> 2017-07-19T12:57:03.176.010.612 (XO: 21.1761523)
   Min dTM/dt [1/us]::-1.00e+03 at an offset of 107.488 usec -> 2017-07-19T12:57:03.176.007.237 (XO: 21.1761490)

 - Block number: 3 starting at 2017-07-19T12:57:21.986.978.750 (XO: 39.9871205, t=34.0°C)  Rank: 00005
   TM values: Max =    78  Min =  -126  Ave =    -4.49  Std =     6.72
   Min TM value:-126 at an offset of 260.114 usec -> 2017-07-19T12:57:21.987.238.863 (XO: 39.9873807)
   Max TM value:78 at an offset of 260.214 usec -> 2017-07-19T12:57:21.987.238.963 (XO: 39.9873808)
   Max dTM/dt [1/us]::2.00e+03 at an offset of 260.164 usec -> 2017-07-19T12:57:21.987.238.913 (XO: 39.9873807)
   Min dTM/dt [1/us]::-9.30e+02 at an offset of 260.064 usec -> 2017-07-19T12:57:21.987.238.813 (XO: 39.9873806)
