-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170719_062311_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 9.131886 to 39.385346
-each separately from 9s to 39s and from 131886us to 385346us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.026806 in 1 packets - #/No: 18/18
      24.225787 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-19T06:23:23.131.886.000 to 2017-07-19T06:23:53.385.346.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.026806 in 1 packets - #/No: 18/18
      24.225787 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4383519 to 40.4382378
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.9999771 in 1 packets - #/No: 18/18
      23.9999088 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170719_062311_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5702390 to 10.5705280
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.0267830  -- OK for SL --
   MIU time from 2017-07-19T06:23:23.131.886.000 to 2017-07-19T06:23:23.132.175.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.026806

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5973110 to 16.5976000
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.2256950  -- OK for SL --
   MIU time from 2017-07-19T06:23:29.158.981.000 to 2017-07-19T06:23:29.159.270.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.225787

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.8232950 to 40.8235840
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-19T06:23:53.385.057.000 to 2017-07-19T06:23:53.385.346.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-19T06:23:23.131.745.750
   Max SL corrected packet time: 2017-07-19T06:23:53.385.205.750
   Min SL corrected XO rel time: 9.1318876
   Max SL corrected XO rel time: 39.3853476
   TM values from -44 to 35 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-19T06:23:23.131.745.750 (XO: 9.1318876, t=34.0°C)  Rank: 00012
   TM values: Max =    35  Min =   -44  Ave =    -4.42  Std =    11.06
   Min TM value:-44 at an offset of 123.888 usec -> 2017-07-19T06:23:23.131.869.637 (XO: 9.1320115)
   Max TM value:35 at an offset of 123.425 usec -> 2017-07-19T06:23:23.131.869.175 (XO: 9.1320110)
   Max dTM/dt [1/us]::1.76e+02 at an offset of 151.688 usec -> 2017-07-19T06:23:23.131.897.438 (XO: 9.1320393)
   Min dTM/dt [1/us]::-1.77e+02 at an offset of 88.700 usec -> 2017-07-19T06:23:23.131.834.450 (XO: 9.1319763)

 - Block number: 2 starting at 2017-07-19T06:23:29.158.840.750 (XO: 15.1589826, t=34.0°C)  Rank: 00009
   TM values: Max =    32  Min =   -41  Ave =    -4.51  Std =    11.19
   Min TM value:-41 at an offset of 147.463 usec -> 2017-07-19T06:23:29.158.988.213 (XO: 15.1591301)
   Max TM value:32 at an offset of 159.288 usec -> 2017-07-19T06:23:29.159.000.038 (XO: 15.1591419)
   Max dTM/dt [1/us]::2.30e+02 at an offset of 305.989 usec -> 2017-07-19T06:23:29.159.146.738 (XO: 15.1592886)
   Min dTM/dt [1/us]::-1.69e+02 at an offset of 111.000 usec -> 2017-07-19T06:23:29.158.951.750 (XO: 15.1590936)

 - Block number: 3 starting at 2017-07-19T06:23:53.384.916.750 (XO: 39.3850586, t=34.0°C)  Rank: 00010
   TM values: Max =    30  Min =   -35  Ave =    -4.57  Std =     9.77
   Min TM value:-35 at an offset of 111.325 usec -> 2017-07-19T06:23:53.385.028.075 (XO: 39.3851700)
   Max TM value:30 at an offset of 69.338 usec -> 2017-07-19T06:23:53.384.986.087 (XO: 39.3851280)
   Max dTM/dt [1/us]::2.41e+02 at an offset of 85.588 usec -> 2017-07-19T06:23:53.385.002.337 (XO: 39.3851442)
   Min dTM/dt [1/us]::-2.27e+02 at an offset of 305.989 usec -> 2017-07-19T06:23:53.385.222.738 (XO: 39.3853646)
