-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170711_021854_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.004782 to 32.036469
-each separately from 1s to 32s and from 4782us to 965479us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.960119 in 1 packets - #/No: 18/18
      16.070701 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-11T02:18:58.004.782.000 to 2017-07-11T02:19:29.036.469.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.960119 in 1 packets - #/No: 18/18
      16.070701 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9394092 to 33.9392903
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999463 in 1 packets - #/No: 18/18
      16.9999348 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170711_021854_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.9441918 to 2.9444808
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9600630  -- OK for SL --
   MIU time from 2017-07-11T02:18:58.004.782.000 to 2017-07-11T02:18:58.005.071.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.960119

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.9045438 to 17.9048328
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.0706390  -- OK for SL --
   MIU time from 2017-07-11T02:19:12.965.190.000 to 2017-07-11T02:19:12.965.479.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.070701

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.9754718 to 33.9757608
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-11T02:19:29.036.180.000 to 2017-07-11T02:19:29.036.469.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-11T02:18:58.004.641.750
   Max SL corrected packet time: 2017-07-11T02:19:29.036.328.750
   Min SL corrected XO rel time: 1.0047826
   Max SL corrected XO rel time: 32.0364706
   TM values from -30 to 23 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-11T02:18:58.004.641.750 (XO: 1.0047826, t=34.0°C)  Rank: 00004
   TM values: Max =    23  Min =   -30  Ave =    -4.53  Std =     7.29
   Min TM value:-30 at an offset of 96.663 usec -> 2017-07-11T02:18:58.004.738.412 (XO: 1.0048793)
   Max TM value:23 at an offset of 103.400 usec -> 2017-07-11T02:18:58.004.745.150 (XO: 1.0048860)
   Max dTM/dt [1/us]::3.17e+02 at an offset of 15.488 usec -> 2017-07-11T02:18:58.004.657.237 (XO: 1.0047981)
   Min dTM/dt [1/us]::-3.24e+02 at an offset of 163.726 usec -> 2017-07-11T02:18:58.004.805.475 (XO: 1.0049463)

 - Block number: 2 starting at 2017-07-11T02:19:12.965.049.750 (XO: 15.9651921, t=34.0°C)  Rank: 00003
   TM values: Max =    14  Min =   -20  Ave =    -4.42  Std =     4.30
   Min TM value:-20 at an offset of 112.950 usec -> 2017-07-11T02:19:12.965.162.700 (XO: 15.9653050)
   Max TM value:14 at an offset of 21.363 usec -> 2017-07-11T02:19:12.965.071.112 (XO: 15.9652134)
   Max dTM/dt [1/us]::1.77e+02 at an offset of 133.776 usec -> 2017-07-11T02:19:12.965.183.525 (XO: 15.9653258)
   Min dTM/dt [1/us]::-1.86e+02 at an offset of 21.713 usec -> 2017-07-11T02:19:12.965.071.462 (XO: 15.9652138)

 - Block number: 3 starting at 2017-07-11T02:19:29.036.039.750 (XO: 32.0361816, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -21  Ave =    -4.52  Std =     4.17
   Min TM value:-21 at an offset of 122.375 usec -> 2017-07-11T02:19:29.036.162.125 (XO: 32.0363040)
   Max TM value:12 at an offset of 92.225 usec -> 2017-07-11T02:19:29.036.131.975 (XO: 32.0362739)
   Max dTM/dt [1/us]::1.81e+02 at an offset of 122.600 usec -> 2017-07-11T02:19:29.036.162.350 (XO: 32.0363042)
   Min dTM/dt [1/us]::-1.91e+02 at an offset of 173.838 usec -> 2017-07-11T02:19:29.036.213.588 (XO: 32.0363555)
