-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170711_001750_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.327790 to 30.221731
-each separately from 12s to 30s and from 221442us to 504082us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.175714 in 1 packets - #/No: 18/18
      8.717360 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-11T00:18:05.327.790.000 to 2017-07-11T00:18:23.221.731.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.175714 in 1 packets - #/No: 18/18
      8.717360 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4390350 to 31.4389657
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999653 in 1 packets - #/No: 18/18
      8.9999654 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170711_001750_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.7668246 to 13.7671136
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1756790  -- OK for SL --
   MIU time from 2017-07-11T00:18:05.327.790.000 to 2017-07-11T00:18:05.328.079.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.175714

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.9427926 to 22.9430816
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7173270  -- OK for SL --
   MIU time from 2017-07-11T00:18:14.503.793.000 to 2017-07-11T00:18:14.504.082.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.717360

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.6604086 to 31.6606976
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-11T00:18:23.221.442.000 to 2017-07-11T00:18:23.221.731.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-11T00:18:05.327.649.750
   Max SL corrected packet time: 2017-07-11T00:18:23.221.590.750
   Min SL corrected XO rel time: 12.3277908
   Max SL corrected XO rel time: 30.2217328
   TM values from -37 to 29 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-11T00:18:05.327.649.750 (XO: 12.3277908, t=34.0°C)  Rank: 00012
   TM values: Max =    29  Min =   -37  Ave =    -4.53  Std =    11.93
   Min TM value:-37 at an offset of 154.813 usec -> 2017-07-11T00:18:05.327.804.563 (XO: 12.3279457)
   Max TM value:29 at an offset of 105.125 usec -> 2017-07-11T00:18:05.327.754.875 (XO: 12.3278960)
   Max dTM/dt [1/us]::4.08e+02 at an offset of 160.988 usec -> 2017-07-11T00:18:05.327.810.738 (XO: 12.3279518)
   Min dTM/dt [1/us]::-4.30e+02 at an offset of 107.338 usec -> 2017-07-11T00:18:05.327.757.087 (XO: 12.3278982)

 - Block number: 2 starting at 2017-07-11T00:18:14.503.652.750 (XO: 21.5037942, t=34.0°C)  Rank: 00008
   TM values: Max =    23  Min =   -31  Ave =    -4.45  Std =     9.07
   Min TM value:-31 at an offset of 119.438 usec -> 2017-07-11T00:18:14.503.772.187 (XO: 21.5039137)
   Max TM value:23 at an offset of 128.363 usec -> 2017-07-11T00:18:14.503.781.112 (XO: 21.5039226)
   Max dTM/dt [1/us]::3.44e+02 at an offset of 146.051 usec -> 2017-07-11T00:18:14.503.798.800 (XO: 21.5039403)
   Min dTM/dt [1/us]::-3.17e+02 at an offset of 120.250 usec -> 2017-07-11T00:18:14.503.773.000 (XO: 21.5039145)

 - Block number: 3 starting at 2017-07-11T00:18:23.221.301.750 (XO: 30.2214438, t=34.0°C)  Rank: 00006
   TM values: Max =    24  Min =   -28  Ave =    -4.53  Std =     7.53
   Min TM value:-28 at an offset of 108.725 usec -> 2017-07-11T00:18:23.221.410.475 (XO: 30.2215525)
   Max TM value:24 at an offset of 129.751 usec -> 2017-07-11T00:18:23.221.431.500 (XO: 30.2215735)
   Max dTM/dt [1/us]::3.85e+02 at an offset of 198.138 usec -> 2017-07-11T00:18:23.221.499.888 (XO: 30.2216419)
   Min dTM/dt [1/us]::-3.64e+02 at an offset of 200.388 usec -> 2017-07-11T00:18:23.221.502.138 (XO: 30.2216442)
