-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170709_142522_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.090612 to 33.990851
-each separately from 1s to 33s and from 90612us to 990851us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.638397 in 1 packets - #/No: 18/18
      8.260975 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-09T14:25:26.090.612.000 to 2017-07-09T14:25:58.990.851.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.638397 in 1 packets - #/No: 18/18
      8.260975 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4374045 to 34.4372814
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999076 in 1 packets - #/No: 18/18
      7.9999693 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170709_142522_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.5280173 to 2.5283063
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.6383030  -- OK for SL --
   MIU time from 2017-07-09T14:25:26.090.612.000 to 2017-07-09T14:25:26.090.901.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.638397

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      23 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.1666093 to 27.1668983
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2609430  -- OK for SL --
   MIU time from 2017-07-09T14:25:50.729.298.000 to 2017-07-09T14:25:50.729.587.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.260975

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.4278413 to 35.4281303
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-09T14:25:58.990.562.000 to 2017-07-09T14:25:58.990.851.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-09T14:25:26.090.471.750
   Max SL corrected packet time: 2017-07-09T14:25:58.990.710.750
   Min SL corrected XO rel time: 1.0906132
   Max SL corrected XO rel time: 33.9908528
   TM values from -42 to 40 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-09T14:25:26.090.471.750 (XO: 1.0906132, t=34.0°C)  Rank: 00004
   TM values: Max =    16  Min =   -25  Ave =    -4.43  Std =     5.87
   Min TM value:-25 at an offset of 106.213 usec -> 2017-07-09T14:25:26.090.577.962 (XO: 1.0907194)
   Max TM value:16 at an offset of 79.638 usec -> 2017-07-09T14:25:26.090.551.387 (XO: 1.0906928)
   Max dTM/dt [1/us]::1.35e+02 at an offset of 79.150 usec -> 2017-07-09T14:25:26.090.550.900 (XO: 1.0906923)
   Min dTM/dt [1/us]::-1.58e+02 at an offset of 58.613 usec -> 2017-07-09T14:25:26.090.530.362 (XO: 1.0906718)

 - Block number: 2 starting at 2017-07-09T14:25:50.729.157.750 (XO: 25.7293000, t=34.0°C)  Rank: 00023
   TM values: Max =    40  Min =   -42  Ave =    -4.40  Std =     9.18
   Min TM value:-42 at an offset of 23.425 usec -> 2017-07-09T14:25:50.729.181.175 (XO: 25.7293234)
   Max TM value:40 at an offset of 67.063 usec -> 2017-07-09T14:25:50.729.224.812 (XO: 25.7293670)
   Max dTM/dt [1/us]::3.83e+02 at an offset of 21.350 usec -> 2017-07-09T14:25:50.729.179.100 (XO: 25.7293213)
   Min dTM/dt [1/us]::-4.05e+02 at an offset of 78.625 usec -> 2017-07-09T14:25:50.729.236.375 (XO: 25.7293786)

 - Block number: 3 starting at 2017-07-09T14:25:58.990.421.750 (XO: 33.9905638, t=34.0°C)  Rank: 00005
   TM values: Max =    20  Min =   -29  Ave =    -4.37  Std =     6.65
   Min TM value:-29 at an offset of 105.613 usec -> 2017-07-09T14:25:58.990.527.362 (XO: 33.9906694)
   Max TM value:20 at an offset of 107.825 usec -> 2017-07-09T14:25:58.990.529.575 (XO: 33.9906716)
   Max dTM/dt [1/us]::1.71e+02 at an offset of 112.088 usec -> 2017-07-09T14:25:58.990.533.837 (XO: 33.9906759)
   Min dTM/dt [1/us]::-1.79e+02 at an offset of 112.563 usec -> 2017-07-09T14:25:58.990.534.312 (XO: 33.9906763)
