-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170704_003813_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.733104 to 38.991846
-each separately from 8s to 38s and from 99681us to 991846us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.366288 in 1 packets - #/No: 18/18
      21.891587 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-04T00:38:24.733.104.000 to 2017-07-04T00:38:54.991.846.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.366288 in 1 packets - #/No: 18/18
      21.891587 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4396749 to 39.4395595
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999653 in 1 packets - #/No: 18/18
      20.9999193 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170704_003813_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.1727779 to 10.1730669
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3662550  -- OK for SL --
   MIU time from 2017-07-04T00:38:24.733.104.000 to 2017-07-04T00:38:24.733.393.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.366288

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.5393219 to 18.5396109
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.8915030  -- OK for SL --
   MIU time from 2017-07-04T00:38:33.099.681.000 to 2017-07-04T00:38:33.099.970.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.891587

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.4311139 to 40.4314029
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-04T00:38:54.991.557.000 to 2017-07-04T00:38:54.991.846.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-04T00:38:24.732.963.750
   Max SL corrected packet time: 2017-07-04T00:38:54.991.705.750
   Min SL corrected XO rel time: 8.7331058
   Max SL corrected XO rel time: 38.9918472
   TM values from -25 to 19 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-04T00:38:24.732.963.750 (XO: 8.7331058, t=34.0°C)  Rank: 00004
   TM values: Max =    17  Min =   -23  Ave =    -4.39  Std =     5.58
   Min TM value:-23 at an offset of 107.013 usec -> 2017-07-04T00:38:24.733.070.762 (XO: 8.7332128)
   Max TM value:17 at an offset of 180.963 usec -> 2017-07-04T00:38:24.733.144.713 (XO: 8.7332867)
   Max dTM/dt [1/us]::3.09e+02 at an offset of 124.163 usec -> 2017-07-04T00:38:24.733.087.912 (XO: 8.7332299)
   Min dTM/dt [1/us]::-3.17e+02 at an offset of 114.438 usec -> 2017-07-04T00:38:24.733.078.187 (XO: 8.7332202)

 - Block number: 2 starting at 2017-07-04T00:38:33.099.540.750 (XO: 17.0996821, t=34.0°C)  Rank: 00003
   TM values: Max =    17  Min =   -23  Ave =    -4.50  Std =     4.74
   Min TM value:-23 at an offset of 131.013 usec -> 2017-07-04T00:38:33.099.671.762 (XO: 17.0998131)
   Max TM value:17 at an offset of 194.776 usec -> 2017-07-04T00:38:33.099.735.525 (XO: 17.0998769)
   Max dTM/dt [1/us]::2.49e+02 at an offset of 142.501 usec -> 2017-07-04T00:38:33.099.683.250 (XO: 17.0998246)
   Min dTM/dt [1/us]::-2.76e+02 at an offset of 199.013 usec -> 2017-07-04T00:38:33.099.739.763 (XO: 17.0998811)

 - Block number: 3 starting at 2017-07-04T00:38:54.991.416.750 (XO: 38.9915582, t=34.0°C)  Rank: 00004
   TM values: Max =    19  Min =   -25  Ave =    -4.42  Std =     4.86
   Min TM value:-25 at an offset of 113.175 usec -> 2017-07-04T00:38:54.991.529.925 (XO: 38.9916713)
   Max TM value:19 at an offset of 114.263 usec -> 2017-07-04T00:38:54.991.531.012 (XO: 38.9916724)
   Max dTM/dt [1/us]::2.05e+02 at an offset of 230.276 usec -> 2017-07-04T00:38:54.991.647.025 (XO: 38.9917884)
   Min dTM/dt [1/us]::-2.05e+02 at an offset of 161.088 usec -> 2017-07-04T00:38:54.991.577.838 (XO: 38.9917193)
