-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170630_153034_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.895149 to 41.811901
-each separately from 8s to 41s and from 811612us to 895438us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.996816 in 1 packets - #/No: 18/18
      11.919069 in 1 packets - #/No: 36/18
MIU packet time from 2017-06-30T15:30:45.895.149.000 to 2017-06-30T15:31:18.811.901.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.996816 in 1 packets - #/No: 18/18
      11.919069 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4384453 to 42.4383179
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999189 in 1 packets - #/No: 18/18
      11.9999537 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170630_153034_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      16 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.3335926 to 10.3338816
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9967350  -- OK for SL --
   MIU time from 2017-06-30T15:30:45.895.149.000 to 2017-06-30T15:30:45.895.438.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.996816

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      21 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.3306166 to 31.3309056
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9190230  -- OK for SL --
   MIU time from 2017-06-30T15:31:06.892.254.000 to 2017-06-30T15:31:06.892.543.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.919069

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.2499286 to 43.2502176
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-06-30T15:31:18.811.612.000 to 2017-06-30T15:31:18.811.901.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-06-30T15:30:45.895.008.750
   Max SL corrected packet time: 2017-06-30T15:31:18.811.760.750
   Min SL corrected XO rel time: 8.8951507
   Max SL corrected XO rel time: 41.8119029
   TM values from -135 to 84 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-06-30T15:30:45.895.008.750 (XO: 8.8951507, t=34.0°C)  Rank: 00016
   TM values: Max =    74  Min =   -54  Ave =    -4.41  Std =     9.69
   Min TM value:-54 at an offset of 100.413 usec -> 2017-06-30T15:30:45.895.109.162 (XO: 8.8952511)
   Max TM value:74 at an offset of 199.788 usec -> 2017-06-30T15:30:45.895.208.538 (XO: 8.8953505)
   Max dTM/dt [1/us]::6.04e+02 at an offset of 176.501 usec -> 2017-06-30T15:30:45.895.185.250 (XO: 8.8953272)
   Min dTM/dt [1/us]::-9.02e+02 at an offset of 176.601 usec -> 2017-06-30T15:30:45.895.185.350 (XO: 8.8953273)

 - Block number: 2 starting at 2017-06-30T15:31:06.892.113.750 (XO: 29.8922559, t=34.0°C)  Rank: 00021
   TM values: Max =    49  Min =  -127  Ave =    -4.51  Std =    10.75
   Min TM value:-127 at an offset of 152.688 usec -> 2017-06-30T15:31:06.892.266.438 (XO: 29.8924086)
   Max TM value:49 at an offset of 129.251 usec -> 2017-06-30T15:31:06.892.243.000 (XO: 29.8923851)
   Max dTM/dt [1/us]::1.41e+03 at an offset of 172.976 usec -> 2017-06-30T15:31:06.892.286.725 (XO: 29.8924289)
   Min dTM/dt [1/us]::-9.26e+02 at an offset of 187.301 usec -> 2017-06-30T15:31:06.892.301.050 (XO: 29.8924432)

 - Block number: 3 starting at 2017-06-30T15:31:18.811.471.750 (XO: 41.8116139, t=34.0°C)  Rank: 00015
   TM values: Max =    84  Min =  -135  Ave =    -4.41  Std =    13.53
   Min TM value:-135 at an offset of 228.301 usec -> 2017-06-30T15:31:18.811.700.050 (XO: 41.8118422)
   Max TM value:84 at an offset of 99.588 usec -> 2017-06-30T15:31:18.811.571.337 (XO: 41.8117135)
   Max dTM/dt [1/us]::1.26e+03 at an offset of 191.676 usec -> 2017-06-30T15:31:18.811.663.425 (XO: 41.8118055)
   Min dTM/dt [1/us]::-1.13e+03 at an offset of 228.201 usec -> 2017-06-30T15:31:18.811.699.950 (XO: 41.8118421)
