-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170624_055921_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.354378 to 41.800091
-each separately from 8s to 41s and from 78808us to 800091us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.724141 in 1 packets - #/No: 18/18
      25.720705 in 1 packets - #/No: 36/18
MIU packet time from 2017-06-24T05:59:32.354.378.000 to 2017-06-24T06:00:05.800.091.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.724141 in 1 packets - #/No: 18/18
      25.720705 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4353739 to 42.4352474
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999694 in 1 packets - #/No: 18/18
      24.9999041 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170624_055921_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.7897521 to 9.7900411
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7241110  -- OK for SL --
   MIU time from 2017-06-24T05:59:32.354.378.000 to 2017-06-24T05:59:32.354.667.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.724141

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.5141521 to 17.5144411
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.7206070  -- OK for SL --
   MIU time from 2017-06-24T05:59:40.078.808.000 to 2017-06-24T05:59:40.079.097.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.720705

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      21 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.2350481 to 43.2353371
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-06-24T06:00:05.799.802.000 to 2017-06-24T06:00:05.800.091.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-06-24T05:59:32.354.237.750
   Max SL corrected packet time: 2017-06-24T06:00:05.799.950.750
   Min SL corrected XO rel time: 8.3543795
   Max SL corrected XO rel time: 41.8000927
   TM values from -103 to 84 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-06-24T05:59:32.354.237.750 (XO: 8.3543795, t=34.0°C)  Rank: 00011
   TM values: Max =    40  Min =   -52  Ave =    -4.36  Std =    14.56
   Min TM value:-52 at an offset of 109.425 usec -> 2017-06-24T05:59:32.354.347.175 (XO: 8.3544890)
   Max TM value:40 at an offset of 108.538 usec -> 2017-06-24T05:59:32.354.346.287 (XO: 8.3544881)
   Max dTM/dt [1/us]::1.78e+02 at an offset of 29.025 usec -> 2017-06-24T05:59:32.354.266.775 (XO: 8.3544086)
   Min dTM/dt [1/us]::-1.66e+02 at an offset of 132.451 usec -> 2017-06-24T05:59:32.354.370.200 (XO: 8.3545120)

 - Block number: 2 starting at 2017-06-24T05:59:40.078.667.750 (XO: 16.0788091, t=34.0°C)  Rank: 00011
   TM values: Max =    28  Min =   -38  Ave =    -4.36  Std =    11.04
   Min TM value:-38 at an offset of 133.476 usec -> 2017-06-24T05:59:40.078.801.225 (XO: 16.0789426)
   Max TM value:28 at an offset of 85.638 usec -> 2017-06-24T05:59:40.078.753.387 (XO: 16.0788947)
   Max dTM/dt [1/us]::2.35e+02 at an offset of 0.013 usec -> 2017-06-24T05:59:40.078.667.762 (XO: 16.0788091)
   Min dTM/dt [1/us]::-1.63e+02 at an offset of 110.400 usec -> 2017-06-24T05:59:40.078.778.150 (XO: 16.0789195)

 - Block number: 3 starting at 2017-06-24T06:00:05.799.661.750 (XO: 41.7998037, t=34.0°C)  Rank: 00021
   TM values: Max =    84  Min =  -103  Ave =    -4.42  Std =    27.49
   Min TM value:-103 at an offset of 129.638 usec -> 2017-06-24T06:00:05.799.791.387 (XO: 41.7999334)
   Max TM value:84 at an offset of 130.113 usec -> 2017-06-24T06:00:05.799.791.862 (XO: 41.7999339)
   Max dTM/dt [1/us]::5.15e+02 at an offset of 136.013 usec -> 2017-06-24T06:00:05.799.797.763 (XO: 41.7999398)
   Min dTM/dt [1/us]::-5.20e+02 at an offset of 142.313 usec -> 2017-06-24T06:00:05.799.804.063 (XO: 41.7999461)
