-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170618_135240_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.899697 to 35.979203
-each separately from 1s to 35s and from 834784us to 979203us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.934798 in 1 packets - #/No: 18/18
      13.143841 in 1 packets - #/No: 36/18
MIU packet time from 2017-06-18T13:52:44.899.697.000 to 2017-06-18T13:53:18.979.203.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.934798 in 1 packets - #/No: 18/18
      13.143841 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4384673 to 36.4383391
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999208 in 1 packets - #/No: 18/18
      12.9999510 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170618_135240_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3381628 to 3.3384518
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9347190  -- OK for SL --
   MIU time from 2017-06-18T13:52:44.899.697.000 to 2017-06-18T13:52:44.899.986.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.934798

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2731708 to 24.2734598
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1437910  -- OK for SL --
   MIU time from 2017-06-18T13:53:05.834.784.000 to 2017-06-18T13:53:05.835.073.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.143841

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      38 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.4172508 to 37.4175398
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-06-18T13:53:18.978.914.000 to 2017-06-18T13:53:18.979.203.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-06-18T13:52:44.899.556.750
   Max SL corrected packet time: 2017-06-18T13:53:18.979.062.750
   Min SL corrected XO rel time: 1.8996989
   Max SL corrected XO rel time: 35.9792044
   TM values from -161 to 156 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-06-18T13:52:44.899.556.750 (XO: 1.8996989, t=34.0°C)  Rank: 00010
   TM values: Max =    17  Min =   -27  Ave =    -4.37  Std =     4.77
   Min TM value:-27 at an offset of 92.450 usec -> 2017-06-18T13:52:44.899.649.200 (XO: 1.8997914)
   Max TM value:17 at an offset of 96.663 usec -> 2017-06-18T13:52:44.899.653.412 (XO: 1.8997956)
   Max dTM/dt [1/us]::8.00e+01 at an offset of 57.838 usec -> 2017-06-18T13:52:44.899.614.587 (XO: 1.8997568)
   Min dTM/dt [1/us]::-8.10e+01 at an offset of 96.888 usec -> 2017-06-18T13:52:44.899.653.637 (XO: 1.8997958)

 - Block number: 2 starting at 2017-06-18T13:53:05.834.643.750 (XO: 22.8347859, t=34.0°C)  Rank: 00009
   TM values: Max =    16  Min =   -26  Ave =    -4.33  Std =     5.22
   Min TM value:-26 at an offset of 128.813 usec -> 2017-06-18T13:53:05.834.772.562 (XO: 22.8349147)
   Max TM value:16 at an offset of 132.901 usec -> 2017-06-18T13:53:05.834.776.650 (XO: 22.8349188)
   Max dTM/dt [1/us]::8.00e+01 at an offset of 129.513 usec -> 2017-06-18T13:53:05.834.773.262 (XO: 22.8349154)
   Min dTM/dt [1/us]::-8.60e+01 at an offset of 6.775 usec -> 2017-06-18T13:53:05.834.650.525 (XO: 22.8347926)

 - Block number: 3 starting at 2017-06-18T13:53:18.978.773.750 (XO: 35.9789154, t=34.0°C)  Rank: 00038
   TM values: Max =   156  Min =  -161  Ave =    -4.34  Std =    91.00
   Min TM value:-161 at an offset of 234.801 usec -> 2017-06-18T13:53:18.979.008.550 (XO: 35.9791502)
   Max TM value:156 at an offset of 235.926 usec -> 2017-06-18T13:53:18.979.009.675 (XO: 35.9791513)
   Max dTM/dt [1/us]::8.62e+02 at an offset of 284.026 usec -> 2017-06-18T13:53:18.979.057.776 (XO: 35.9791994)
   Min dTM/dt [1/us]::-1.02e+03 at an offset of 0.000 usec -> 2017-06-18T13:53:18.978.773.750 (XO: 35.9789154)
