-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170617_152523_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.098200 to 30.429848
-each separately from 0s to 30s and from 98200us to 742706us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.643928 in 1 packets - #/No: 18/18
      14.686853 in 1 packets - #/No: 36/18
MIU packet time from 2017-06-17T15:25:26.098.200.000 to 2017-06-17T15:25:56.429.848.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.643928 in 1 packets - #/No: 18/18
      14.686853 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4461989 to 31.4460890
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999451 in 1 packets - #/No: 18/18
      14.9999450 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170617_152523_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5443995 to 1.5446885
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6438710  -- OK for SL --
   MIU time from 2017-06-17T15:25:26.098.200.000 to 2017-06-17T15:25:26.098.489.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.643928

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1885595 to 17.1888485
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6867990  -- OK for SL --
   MIU time from 2017-06-17T15:25:41.742.417.000 to 2017-06-17T15:25:41.742.706.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.686853

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.8756475 to 31.8759365
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-06-17T15:25:56.429.559.000 to 2017-06-17T15:25:56.429.848.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-06-17T15:25:26.098.059.750
   Max SL corrected packet time: 2017-06-17T15:25:56.429.707.750
   Min SL corrected XO rel time: 0.0982010
   Max SL corrected XO rel time: 30.4298491
   TM values from -25 to 18 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-06-17T15:25:26.098.059.750 (XO: 0.0982010, t=34.0°C)  Rank: 00005
   TM values: Max =    14  Min =   -23  Ave =    -4.55  Std =     5.34
   Min TM value:-23 at an offset of 274.126 usec -> 2017-06-17T15:25:26.098.333.876 (XO: 0.0984751)
   Max TM value:14 at an offset of 113.363 usec -> 2017-06-17T15:25:26.098.173.112 (XO: 0.0983143)
   Max dTM/dt [1/us]::1.17e+02 at an offset of 70.475 usec -> 2017-06-17T15:25:26.098.130.225 (XO: 0.0982714)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 203.713 usec -> 2017-06-17T15:25:26.098.263.463 (XO: 0.0984047)

 - Block number: 2 starting at 2017-06-17T15:25:41.742.276.750 (XO: 15.7424182, t=34.0°C)  Rank: 00005
   TM values: Max =    14  Min =   -21  Ave =    -4.52  Std =     4.26
   Min TM value:-21 at an offset of 161.888 usec -> 2017-06-17T15:25:41.742.438.638 (XO: 15.7425801)
   Max TM value:14 at an offset of 92.788 usec -> 2017-06-17T15:25:41.742.369.537 (XO: 15.7425110)
   Max dTM/dt [1/us]::8.60e+01 at an offset of 215.388 usec -> 2017-06-17T15:25:41.742.492.138 (XO: 15.7426336)
   Min dTM/dt [1/us]::-8.70e+01 at an offset of 79.238 usec -> 2017-06-17T15:25:41.742.355.987 (XO: 15.7424974)

 - Block number: 3 starting at 2017-06-17T15:25:56.429.418.750 (XO: 30.4295601, t=34.0°C)  Rank: 00005
   TM values: Max =    18  Min =   -25  Ave =    -4.51  Std =     5.50
   Min TM value:-25 at an offset of 115.900 usec -> 2017-06-17T15:25:56.429.534.650 (XO: 30.4296760)
   Max TM value:18 at an offset of 106.025 usec -> 2017-06-17T15:25:56.429.524.775 (XO: 30.4296661)
   Max dTM/dt [1/us]::1.14e+02 at an offset of 105.775 usec -> 2017-06-17T15:25:56.429.524.525 (XO: 30.4296659)
   Min dTM/dt [1/us]::-1.22e+02 at an offset of 261.601 usec -> 2017-06-17T15:25:56.429.680.350 (XO: 30.4298217)
