-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170615_054952_1229.dat, 36720  packets found.
APID:
      1229 (EW) in 36720 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36720 packets (all packets)
Packet no from 1 to 36720
   Differences of packet numbers in neigboring packets: 
      1 in 36719 packets (all packets)
Total number of packets: 
      36720 in 36720 packets (all packets)
Hz and MHz counters from 31.178945 to 32.129181
-each separately from 31s to 32s and from 129181us to 494534us.
   Differences in neigboring packets: 
      0.000000 in 36716 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -0.950236 in 1 packets - #/No: 8772/8772
      0.315589 in 1 packets - #/No: 21012/21012
      0.634647 in 1 packets - #/No: 33252/33252
MIU packet time from 2017-06-15T05:50:26.178.945.000 to 2017-06-15T05:50:27.129.181.000
   Differences in neigboring packets: 
      0.000000 in 36716 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -0.950236 in 1 packets - #/No: 8772/8772
      0.315589 in 1 packets - #/No: 21012/21012
      0.634647 in 1 packets - #/No: 33252/33252
MIU orbit number: 
      60000 in 36720 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36720 packets (all packets)
MIU orbit information / number of events: 
      0 in 36720 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36720 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36720 packets (all packets)
FPGA configuration version: 
      211 in 36720 packets (all packets)
10 MHz counter at the last 1Hz pulse from 32.5744579 to 33.5744541
   Differences in neigboring packets: 
      0.0000000 in 36717 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -0.9999962 in 1 packets - #/No: 8772/8772
      0.9999962 in 1 packets - #/No: 33252/33252
Temperature FPGA: 
      07 -> 45.9°C in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 24480 packets - #/No: 8773/8773, 8774/8774, 8775/8775, 8776/8776, 8777/8777, 8778/8778, 8779/8779, 8780/8780, 8781/8781, 8782/8782, 8783/8783, 8784/8784, 8785/8785, 8786/8786, 8787/8787, 8788/8788, 8789/8789, 8790/8790, ...
Temperature XO: 
      BF -> 34.0°C in 36720 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 36720 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 36720 packets (all packets)
Number of external events: 
      3 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 12240 packets - #/No: 8773/8773, 8774/8774, 8775/8775, 8776/8776, 8777/8777, 8778/8778, 8779/8779, 8780/8780, 8781/8781, 8782/8782, 8783/8783, 8784/8784, 8785/8785, 8786/8786, 8787/8787, 8788/8788, 8789/8789, 8790/8790, ...
      2 in 12240 packets - #/No: 21013/21013, 21014/21014, 21015/21015, 21016/21016, 21017/21017, 21018/21018, 21019/21019, 21020/21020, 21021/21021, 21022/21022, 21023/21023, 21024/21024, 21025/21025, 21026/21026, 21027/21027, 21028/21028, 21029/21029, 21030/21030, ...
Number of generated alerts: 
      3 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 12240 packets - #/No: 8773/8773, 8774/8774, 8775/8775, 8776/8776, 8777/8777, 8778/8778, 8779/8779, 8780/8780, 8781/8781, 8782/8782, 8783/8783, 8784/8784, 8785/8785, 8786/8786, 8787/8787, 8788/8788, 8789/8789, 8790/8790, ...
      2 in 12240 packets - #/No: 21013/21013, 21014/21014, 21015/21015, 21016/21016, 21017/21017, 21018/21018, 21019/21019, 21020/21020, 21021/21021, 21022/21022, 21023/21023, 21024/21024, 21025/21025, 21026/21026, 21027/21027, 21028/21028, 21029/21029, 21030/21030, ...
Survey memory error: 
      0 in 36720 packets (all packets)
Event memory error: 
      0 in 36720 packets (all packets)
Configuration error: 
      0 in 36720 packets (all packets)
Zero: 
      0 in 36720 packets (all packets)
Radio gain: 
      0 in 36720 packets (all packets)
Radio bandwidth: 
      0 in 36720 packets (all packets)
Digital gain: 
      0 in 36720 packets (all packets)
Analog gain: 
      0 in 36720 packets (all packets)

EVENT DATA READER
-- File 20170615_054952_1229.dat, 180 EW pages found. Extra packets: 0.
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      3 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 12240 packets - #/No: 8773/8773, 8774/8774, 8775/8775, 8776/8776, 8777/8777, 8778/8778, 8779/8779, 8780/8780, 8781/8781, 8782/8782, 8783/8783, 8784/8784, 8785/8785, 8786/8786, 8787/8787, 8788/8788, 8789/8789, 8790/8790, ...
      2 in 12240 packets - #/No: 21013/21013, 21014/21014, 21015/21015, 21016/21016, 21017/21017, 21018/21018, 21019/21019, 21020/21020, 21021/21021, 21022/21022, 21023/21023, 21024/21024, 21025/21025, 21026/21026, 21027/21027, 21028/21028, 21029/21029, 21030/21030, ...
   10 MHz counter from 32.6979153 to 33.8562103
      Difference between the neighboring packets: 
      0.0000170 in 36716 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -1.1582950 in 1 packets - #/No: 8772/8772
      0.1075250 in 1 packets - #/No: 21012/21012
      0.4265810 in 1 packets - #/No: 33252/33252
   Memory page number from 0 to 237
      Difference between the neighboring packets: 
      0 in 36540 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 176 packets - #/No: 204/204, 408/408, 612/612, 816/816, 1020/1020, 1224/1224, 1428/1428, 1632/1632, 1836/1836, 2040/2040, 2244/2244, 2448/2448, 2652/2652, 2856/2856, 3060/3060, 3264/3264, 3468/3468, 3672/3672, ...
      23 in 1 packets - #/No: 8772/8772
      32 in 1 packets - #/No: 21012/21012
      6 in 1 packets - #/No: 33252/33252
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 36540 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 176 packets - #/No: 204/204, 408/408, 612/612, 816/816, 1020/1020, 1224/1224, 1428/1428, 1632/1632, 1836/1836, 2040/2040, 2244/2244, 2448/2448, 2652/2652, 2856/2856, 3060/3060, 3264/3264, 3468/3468, 3672/3672, ...
      -59 in 3 packets - #/No: 8772/8772, 21012/21012, 33252/33252
   MIU event flag 
      1 in 36720 packets (all packets)
   Time triggered event flag 
      0 in 36720 packets (all packets)
   Min EW event (1st sample) abs. time: 2017-06-15T05:50:26.123.456.540
   Min EW header abs. time            : 2017-06-15T05:50:26.178.945.000
   Max EW event (1st sample) abs. time: 2017-06-15T05:50:27.073.692.540
   Max EW header abs. time            : 2017-06-15T05:50:27.129.181.000
   TM values from -164 to 146 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      106 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      107 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      108 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      109 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      110 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      111 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      112 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      113 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      114 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      115 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      116 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      117 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      118 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      119 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      120 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      121 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      122 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      123 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      124 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 60  ordered packets: 12240
   TM values from -164 to 146 median:-4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=31 C1MHz=0.178945
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=32.6979153
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2017-06-15T05:50:26.123.456.540
      (header abs. time: 2017-06-15T05:50:26.178.945.000)
      Event 1st sample XO time (corr. sec. from XP start):              31.123457319
      Event header XO time (corr. sec. from XP start)    :              31.123457569
      Sampling frequency correction factor for this event:     0.999996200

 Event no. 2 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      156 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      157 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      158 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      159 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      160 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      161 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      162 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      163 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      164 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      165 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      166 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      167 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      168 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      169 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      170 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      171 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      172 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      173 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      174 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      175 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      176 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      177 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      178 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      179 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      180 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      181 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      182 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      183 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      184 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      185 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      186 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      187 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      188 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      189 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      190 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      191 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      192 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      193 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      194 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      195 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      196 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      197 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      198 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      199 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      200 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      201 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      202 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      203 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      204 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      205 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      206 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      207 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      208 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      209 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      210 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      211 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      212 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      213 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      214 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      215 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 60  ordered packets: 12240
   TM values from -144 to 139 median:-4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=31 C1MHz=0.494534
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=33.0135033
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2017-06-15T05:50:26.439.045.540
      (header abs. time: 2017-06-15T05:50:26.494.534.000)
      Event 1st sample XO time (corr. sec. from XP start):              31.439046518
      Event header XO time (corr. sec. from XP start)    :              31.439046768
      Sampling frequency correction factor for this event:     0.999996200

 Event no. 3 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      221 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      222 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      223 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      224 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      225 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      226 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      227 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      228 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      229 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      230 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      231 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      232 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      233 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      234 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      235 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      236 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      237 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 60  ordered packets: 12240
   TM values from -155 to 140 median:-4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=32 C1MHz=0.129181
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=33.6481473
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2017-06-15T05:50:27.073.692.540
      (header abs. time: 2017-06-15T05:50:27.129.181.000)
      Event 1st sample XO time (corr. sec. from XP start):              32.073692930
      Event header XO time (corr. sec. from XP start)    :              32.073693180
      Sampling frequency correction factor for this event:     0.999996200
--
Extremes in separate events:

 - Event number: 1 starting at 2017-06-15T05:50:26.123.456.540 (XO: 31.1234573)
   Min TM value:-164 at an offset of 89887.179 usec -> 2017-06-15T05:50:26.213.343.719 (XO: 31.2133445)
   Max TM value:146 at an offset of 28571.134 usec -> 2017-06-15T05:50:26.152.027.673 (XO: 31.1520285)
   Max dTM/dt [1/us]:8.70e+02 at an offset of 89871.154 usec -> 2017-06-15T05:50:26.213.327.694 (XO: 31.2133285)
   Min dTM/dt [1/us]:-8.63e+02 at an offset of 89880.654 usec -> 2017-06-15T05:50:26.213.337.194 (XO: 31.2133380)

 - Event number: 2 starting at 2017-06-15T05:50:26.439.045.540 (XO: 31.4390465)
   Min TM value:-144 at an offset of 184769.590 usec -> 2017-06-15T05:50:26.623.815.129 (XO: 31.6238161)
   Max TM value:139 at an offset of 184793.565 usec -> 2017-06-15T05:50:26.623.839.104 (XO: 31.6238401)
   Max dTM/dt [1/us]:8.33e+02 at an offset of 184795.840 usec -> 2017-06-15T05:50:26.623.841.379 (XO: 31.6238424)
   Min dTM/dt [1/us]:-8.77e+02 at an offset of 184787.290 usec -> 2017-06-15T05:50:26.623.832.829 (XO: 31.6238338)

 - Event number: 3 starting at 2017-06-15T05:50:27.073.692.540 (XO: 32.0736929)
   Min TM value:-155 at an offset of 143632.983 usec -> 2017-06-15T05:50:27.217.325.523 (XO: 32.2173259)
   Max TM value:140 at an offset of 143579.021 usec -> 2017-06-15T05:50:27.217.271.560 (XO: 32.2172720)
   Max dTM/dt [1/us]:8.34e+02 at an offset of 143671.396 usec -> 2017-06-15T05:50:27.217.363.935 (XO: 32.2173643)
   Min dTM/dt [1/us]:-8.09e+02 at an offset of 143544.433 usec -> 2017-06-15T05:50:27.217.236.972 (XO: 32.2172374)
