-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170606_084244_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.338432 to 35.887046
-each separately from 5s to 35s and from 48341us to 887046us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.709620 in 1 packets - #/No: 18/18
      20.838127 in 1 packets - #/No: 36/18
MIU packet time from 2017-06-06T08:42:52.338.432.000 to 2017-06-06T08:43:22.887.046.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.709620 in 1 packets - #/No: 18/18
      20.838127 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4366475 to 36.4365319
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999615 in 1 packets - #/No: 18/18
      19.9999229 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170606_084244_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.7750793 to 6.7753683
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.7095830  -- OK for SL --
   MIU time from 2017-06-06T08:42:52.338.432.000 to 2017-06-06T08:42:52.338.721.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.709620

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4849513 to 16.4852403
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.8380470  -- OK for SL --
   MIU time from 2017-06-06T08:43:02.048.341.000 to 2017-06-06T08:43:02.048.630.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.838127

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.3232873 to 37.3235763
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-06-06T08:43:22.886.757.000 to 2017-06-06T08:43:22.887.046.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-06-06T08:42:52.338.291.750
   Max SL corrected packet time: 2017-06-06T08:43:22.886.905.750
   Min SL corrected XO rel time: 5.3384331
   Max SL corrected XO rel time: 35.8870479
   TM values from -58 to 43 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-06-06T08:42:52.338.291.750 (XO: 5.3384331, t=35.6°C)  Rank: 00011
   TM values: Max =    43  Min =   -58  Ave =    -4.40  Std =    17.00
   Min TM value:-58 at an offset of 119.150 usec -> 2017-06-06T08:42:52.338.410.900 (XO: 5.3385522)
   Max TM value:43 at an offset of 118.213 usec -> 2017-06-06T08:42:52.338.409.962 (XO: 5.3385513)
   Max dTM/dt [1/us]::3.71e+02 at an offset of 152.901 usec -> 2017-06-06T08:42:52.338.444.650 (XO: 5.3385860)
   Min dTM/dt [1/us]::-3.80e+02 at an offset of 118.650 usec -> 2017-06-06T08:42:52.338.410.400 (XO: 5.3385517)

 - Block number: 2 starting at 2017-06-06T08:43:02.048.200.750 (XO: 15.0483425, t=35.6°C)  Rank: 00007
   TM values: Max =    25  Min =   -34  Ave =    -4.47  Std =     7.95
   Min TM value:-34 at an offset of 99.988 usec -> 2017-06-06T08:43:02.048.300.737 (XO: 15.0484425)
   Max TM value:25 at an offset of 100.813 usec -> 2017-06-06T08:43:02.048.301.562 (XO: 15.0484433)
   Max dTM/dt [1/us]::1.96e+02 at an offset of 100.763 usec -> 2017-06-06T08:43:02.048.301.512 (XO: 15.0484433)
   Min dTM/dt [1/us]::-1.94e+02 at an offset of 96.350 usec -> 2017-06-06T08:43:02.048.297.100 (XO: 15.0484388)

 - Block number: 3 starting at 2017-06-06T08:43:22.886.616.750 (XO: 35.8867589, t=35.6°C)  Rank: 00008
   TM values: Max =    28  Min =   -36  Ave =    -4.41  Std =     8.88
   Min TM value:-36 at an offset of 122.438 usec -> 2017-06-06T08:43:22.886.739.187 (XO: 35.8868813)
   Max TM value:28 at an offset of 118.088 usec -> 2017-06-06T08:43:22.886.734.837 (XO: 35.8868769)
   Max dTM/dt [1/us]::2.47e+02 at an offset of 107.388 usec -> 2017-06-06T08:43:22.886.724.137 (XO: 35.8868662)
   Min dTM/dt [1/us]::-2.06e+02 at an offset of 121.938 usec -> 2017-06-06T08:43:22.886.738.687 (XO: 35.8868808)
