-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170605_093136_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.799982 to 42.734707
-each separately from 3s to 42s and from 597070us to 800271us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.796799 in 1 packets - #/No: 18/18
      22.137059 in 1 packets - #/No: 36/18
MIU packet time from 2017-06-05T09:31:42.799.982.000 to 2017-06-05T09:32:21.734.707.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.796799 in 1 packets - #/No: 18/18
      22.137059 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4491039 to 43.4489556
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999353 in 1 packets - #/No: 18/18
      21.9999164 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170605_093136_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.2490840 to 5.2493730
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7967350  -- OK for SL --
   MIU time from 2017-06-05T09:31:42.799.982.000 to 2017-06-05T09:31:42.800.271.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.796799

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      31 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.0461080 to 22.0463970
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1369750  -- OK for SL --
   MIU time from 2017-06-05T09:31:59.597.070.000 to 2017-06-05T09:31:59.597.359.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.137059

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.1833720 to 44.1836610
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-06-05T09:32:21.734.418.000 to 2017-06-05T09:32:21.734.707.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-06-05T09:31:42.799.841.750
   Max SL corrected packet time: 2017-06-05T09:32:21.734.566.750
   Min SL corrected XO rel time: 3.7999831
   Max SL corrected XO rel time: 42.7347081
   TM values from -77 to 37 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-06-05T09:31:42.799.841.750 (XO: 3.7999831, t=34.0°C)  Rank: 00004
   TM values: Max =    14  Min =   -21  Ave =    -4.42  Std =     5.76
   Min TM value:-21 at an offset of 128.838 usec -> 2017-06-05T09:31:42.799.970.587 (XO: 3.8001120)
   Max TM value:14 at an offset of 228.476 usec -> 2017-06-05T09:31:42.800.070.225 (XO: 3.8002116)
   Max dTM/dt [1/us]::1.17e+02 at an offset of 110.775 usec -> 2017-06-05T09:31:42.799.952.525 (XO: 3.8000939)
   Min dTM/dt [1/us]::-1.12e+02 at an offset of 216.663 usec -> 2017-06-05T09:31:42.800.058.413 (XO: 3.8001998)

 - Block number: 2 starting at 2017-06-05T09:31:59.596.929.750 (XO: 20.5970711, t=34.0°C)  Rank: 00031
   TM values: Max =    37  Min =   -47  Ave =    -4.42  Std =     7.32
   Min TM value:-47 at an offset of 112.250 usec -> 2017-06-05T09:31:59.597.042.000 (XO: 20.5971833)
   Max TM value:37 at an offset of 108.550 usec -> 2017-06-05T09:31:59.597.038.300 (XO: 20.5971796)
   Max dTM/dt [1/us]::8.10e+01 at an offset of 301.526 usec -> 2017-06-05T09:31:59.597.231.276 (XO: 20.5973726)
   Min dTM/dt [1/us]::-8.80e+01 at an offset of 12.913 usec -> 2017-06-05T09:31:59.596.942.662 (XO: 20.5970840)

 - Block number: 3 starting at 2017-06-05T09:32:21.734.277.750 (XO: 42.7344191, t=34.0°C)  Rank: 00013
   TM values: Max =    28  Min =   -77  Ave =    -4.44  Std =     7.67
   Min TM value:-77 at an offset of 120.638 usec -> 2017-06-05T09:32:21.734.398.387 (XO: 42.7345398)
   Max TM value:28 at an offset of 19.113 usec -> 2017-06-05T09:32:21.734.296.862 (XO: 42.7344382)
   Max dTM/dt [1/us]::7.83e+02 at an offset of 62.138 usec -> 2017-06-05T09:32:21.734.339.887 (XO: 42.7344813)
   Min dTM/dt [1/us]::-5.98e+02 at an offset of 62.050 usec -> 2017-06-05T09:32:21.734.339.800 (XO: 42.7344812)
