-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170531_232317_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.065278 to 30.204449
-each separately from 0s to 30s and from 65278us to 396308us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.330452 in 1 packets - #/No: 18/18
      11.807852 in 1 packets - #/No: 36/18
MIU packet time from 2017-05-31T23:23:20.065.278.000 to 2017-05-31T23:23:50.204.449.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.330452 in 1 packets - #/No: 18/18
      11.807852 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4342526 to 31.4341389
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999318 in 1 packets - #/No: 18/18
      11.9999545 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170531_232317_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4995313 to 1.4998203
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3303830  -- OK for SL --
   MIU time from 2017-05-31T23:23:20.065.278.000 to 2017-05-31T23:23:20.065.567.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.330452

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.8302033 to 19.8304923
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8078070  -- OK for SL --
   MIU time from 2017-05-31T23:23:38.396.019.000 to 2017-05-31T23:23:38.396.308.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.807852

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.6382993 to 31.6385883
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-05-31T23:23:50.204.160.000 to 2017-05-31T23:23:50.204.449.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-05-31T23:23:20.065.137.750
   Max SL corrected packet time: 2017-05-31T23:23:50.204.308.750
   Min SL corrected XO rel time: 0.0652789
   Max SL corrected XO rel time: 30.2044502
   TM values from -51 to 36 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-05-31T23:23:20.065.137.750 (XO: 0.0652789, t=34.0°C)  Rank: 00010
   TM values: Max =    34  Min =   -42  Ave =    -4.29  Std =    10.73
   Min TM value:-42 at an offset of 106.850 usec -> 2017-05-31T23:23:20.065.244.600 (XO: 0.0653858)
   Max TM value:34 at an offset of 95.113 usec -> 2017-05-31T23:23:20.065.232.862 (XO: 0.0653741)
   Max dTM/dt [1/us]::5.03e+02 at an offset of 107.713 usec -> 2017-05-31T23:23:20.065.245.462 (XO: 0.0653867)
   Min dTM/dt [1/us]::-5.32e+02 at an offset of 100.988 usec -> 2017-05-31T23:23:20.065.238.737 (XO: 0.0653799)

 - Block number: 2 starting at 2017-05-31T23:23:38.395.878.750 (XO: 18.3960204, t=34.0°C)  Rank: 00010
   TM values: Max =    36  Min =   -51  Ave =    -4.31  Std =    11.83
   Min TM value:-51 at an offset of 94.375 usec -> 2017-05-31T23:23:38.395.973.125 (XO: 18.3961148)
   Max TM value:36 at an offset of 97.900 usec -> 2017-05-31T23:23:38.395.976.650 (XO: 18.3961183)
   Max dTM/dt [1/us]::4.76e+02 at an offset of 303.014 usec -> 2017-05-31T23:23:38.396.181.763 (XO: 18.3963234)
   Min dTM/dt [1/us]::-5.09e+02 at an offset of 267.301 usec -> 2017-05-31T23:23:38.396.146.051 (XO: 18.3962877)

 - Block number: 3 starting at 2017-05-31T23:23:50.204.019.750 (XO: 30.2041612, t=34.0°C)  Rank: 00010
   TM values: Max =    32  Min =   -36  Ave =    -4.32  Std =     8.90
   Min TM value:-36 at an offset of 122.300 usec -> 2017-05-31T23:23:50.204.142.050 (XO: 30.2042835)
   Max TM value:32 at an offset of 125.813 usec -> 2017-05-31T23:23:50.204.145.562 (XO: 30.2042870)
   Max dTM/dt [1/us]::3.75e+02 at an offset of 154.901 usec -> 2017-05-31T23:23:50.204.174.650 (XO: 30.2043161)
   Min dTM/dt [1/us]::-3.64e+02 at an offset of 154.013 usec -> 2017-05-31T23:23:50.204.173.763 (XO: 30.2043152)
