-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170531_061130_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.342375 to 40.891638
-each separately from 8s to 40s and from 342375us to 891638us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.427787 in 1 packets - #/No: 18/18
      17.120609 in 1 packets - #/No: 36/18
MIU packet time from 2017-05-31T06:11:41.342.375.000 to 2017-05-31T06:12:13.891.638.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.427787 in 1 packets - #/No: 18/18
      17.120609 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.1812614 to 42.1811378
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999421 in 1 packets - #/No: 18/18
      16.9999343 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170531_061130_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      38 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5236367 to 10.5239257
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4277270  -- OK for SL --
   MIU time from 2017-05-31T06:11:41.342.375.000 to 2017-05-31T06:11:41.342.664.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.427787

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      27 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.9516527 to 25.9519417
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1205430  -- OK for SL --
   MIU time from 2017-05-31T06:11:56.770.451.000 to 2017-05-31T06:11:56.770.740.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.120609

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      37 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.0724847 to 43.0727737
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-05-31T06:12:13.891.349.000 to 2017-05-31T06:12:13.891.638.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-05-31T06:11:41.342.234.750
   Max SL corrected packet time: 2017-05-31T06:12:13.891.497.750
   Min SL corrected XO rel time: 8.3423766
   Max SL corrected XO rel time: 40.8916394
   TM values from -113 to 104 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-05-31T06:11:41.342.234.750 (XO: 8.3423766, t=34.0°C)  Rank: 00038
   TM values: Max =   100  Min =  -108  Ave =    -4.22  Std =    36.02
   Min TM value:-108 at an offset of 128.276 usec -> 2017-05-31T06:11:41.342.363.025 (XO: 8.3425049)
   Max TM value:100 at an offset of 132.076 usec -> 2017-05-31T06:11:41.342.366.825 (XO: 8.3425087)
   Max dTM/dt [1/us]::4.54e+02 at an offset of 0.000 usec -> 2017-05-31T06:11:41.342.234.750 (XO: 8.3423766)
   Min dTM/dt [1/us]::-4.56e+02 at an offset of 107.125 usec -> 2017-05-31T06:11:41.342.341.875 (XO: 8.3424838)

 - Block number: 2 starting at 2017-05-31T06:11:56.770.310.750 (XO: 23.7704522, t=34.0°C)  Rank: 00027
   TM values: Max =    90  Min =   -96  Ave =    -4.24  Std =    29.87
   Min TM value:-96 at an offset of 113.850 usec -> 2017-05-31T06:11:56.770.424.600 (XO: 23.7705661)
   Max TM value:90 at an offset of 125.075 usec -> 2017-05-31T06:11:56.770.435.825 (XO: 23.7705773)
   Max dTM/dt [1/us]::6.04e+02 at an offset of 174.963 usec -> 2017-05-31T06:11:56.770.485.713 (XO: 23.7706272)
   Min dTM/dt [1/us]::-6.14e+02 at an offset of 305.989 usec -> 2017-05-31T06:11:56.770.616.738 (XO: 23.7707582)

 - Block number: 3 starting at 2017-05-31T06:12:13.891.208.750 (XO: 40.8913504, t=34.0°C)  Rank: 00037
   TM values: Max =   104  Min =  -113  Ave =    -4.37  Std =    39.55
   Min TM value:-113 at an offset of 124.600 usec -> 2017-05-31T06:12:13.891.333.350 (XO: 40.8914750)
   Max TM value:104 at an offset of 149.701 usec -> 2017-05-31T06:12:13.891.358.450 (XO: 40.8915001)
   Max dTM/dt [1/us]::7.38e+02 at an offset of 141.276 usec -> 2017-05-31T06:12:13.891.350.025 (XO: 40.8914917)
   Min dTM/dt [1/us]::-9.27e+02 at an offset of 305.989 usec -> 2017-05-31T06:12:13.891.514.738 (XO: 40.8916564)
