-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170529_161227_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.757654 to 33.540846
-each separately from 2s to 33s and from 239070us to 757943us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.481127 in 1 packets - #/No: 18/18
      12.301198 in 1 packets - #/No: 36/18
MIU packet time from 2017-05-29T16:12:32.757.654.000 to 2017-05-29T16:13:03.540.846.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.481127 in 1 packets - #/No: 18/18
      12.301198 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4374611 to 34.4373415
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.9999267 in 1 packets - #/No: 18/18
      11.9999537 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170529_161227_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      31 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1951137 to 4.1954027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4810550  -- OK for SL --
   MIU time from 2017-05-29T16:12:32.757.654.000 to 2017-05-29T16:12:32.757.943.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.481127

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      49 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.6764577 to 22.6767467
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3011510  -- OK for SL --
   MIU time from 2017-05-29T16:12:51.239.070.000 to 2017-05-29T16:12:51.239.359.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.301198

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      45 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.9778977 to 34.9781867
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-05-29T16:13:03.540.557.000 to 2017-05-29T16:13:03.540.846.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-05-29T16:12:32.757.513.750
   Max SL corrected packet time: 2017-05-29T16:13:03.540.705.750
   Min SL corrected XO rel time: 2.7576556
   Max SL corrected XO rel time: 33.5408473
   TM values from -129 to 121 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-05-29T16:12:32.757.513.750 (XO: 2.7576556, t=34.0°C)  Rank: 00031
   TM values: Max =    82  Min =   -91  Ave =    -4.43  Std =    28.52
   Min TM value:-91 at an offset of 156.526 usec -> 2017-05-29T16:12:32.757.670.275 (XO: 2.7578121)
   Max TM value:82 at an offset of 156.163 usec -> 2017-05-29T16:12:32.757.669.913 (XO: 2.7578117)
   Max dTM/dt [1/us]::6.81e+02 at an offset of 143.426 usec -> 2017-05-29T16:12:32.757.657.175 (XO: 2.7577990)
   Min dTM/dt [1/us]::-6.63e+02 at an offset of 142.676 usec -> 2017-05-29T16:12:32.757.656.425 (XO: 2.7577982)

 - Block number: 2 starting at 2017-05-29T16:12:51.238.929.750 (XO: 21.2390708, t=34.0°C)  Rank: 00049
   TM values: Max =   121  Min =  -129  Ave =    -4.34  Std =    36.27
   Min TM value:-129 at an offset of 118.888 usec -> 2017-05-29T16:12:51.239.048.637 (XO: 21.2391897)
   Max TM value:121 at an offset of 133.576 usec -> 2017-05-29T16:12:51.239.063.325 (XO: 21.2392044)
   Max dTM/dt [1/us]::1.02e+03 at an offset of 134.438 usec -> 2017-05-29T16:12:51.239.064.188 (XO: 21.2392053)
   Min dTM/dt [1/us]::-1.03e+03 at an offset of 134.013 usec -> 2017-05-29T16:12:51.239.063.763 (XO: 21.2392048)

 - Block number: 3 starting at 2017-05-29T16:13:03.540.416.750 (XO: 33.5405583, t=34.0°C)  Rank: 00045
   TM values: Max =    97  Min =  -106  Ave =    -4.43  Std =    35.60
   Min TM value:-106 at an offset of 129.376 usec -> 2017-05-29T16:13:03.540.546.125 (XO: 33.5406877)
   Max TM value:97 at an offset of 143.663 usec -> 2017-05-29T16:13:03.540.560.413 (XO: 33.5407020)
   Max dTM/dt [1/us]::8.37e+02 at an offset of 144.076 usec -> 2017-05-29T16:13:03.540.560.825 (XO: 33.5407024)
   Min dTM/dt [1/us]::-8.35e+02 at an offset of 143.713 usec -> 2017-05-29T16:13:03.540.560.463 (XO: 33.5407020)
