-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170518_075808_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.380203 to 42.209388
-each separately from 0s to 42s and from 209099us to 380492us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.939261 in 1 packets - #/No: 18/18
      12.889057 in 1 packets - #/No: 36/18
MIU packet time from 2017-05-18T07:58:11.380.203.000 to 2017-05-18T07:58:53.209.388.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.939261 in 1 packets - #/No: 18/18
      12.889057 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4307822 to 43.4306220
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.9998895 in 1 packets - #/No: 18/18
      12.9999503 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170518_075808_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8109851 to 1.8112741
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.9391510  -- OK for SL --
   MIU time from 2017-05-18T07:58:11.380.203.000 to 2017-05-18T07:58:11.380.492.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.939261

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.7504251 to 30.7507141
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8890070  -- OK for SL --
   MIU time from 2017-05-18T07:58:40.319.753.000 to 2017-05-18T07:58:40.320.042.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.889057

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      18 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.6397211 to 43.6400101
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-05-18T07:58:53.209.099.000 to 2017-05-18T07:58:53.209.388.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-05-18T07:58:11.380.062.750
   Max SL corrected packet time: 2017-05-18T07:58:53.209.247.750
   Min SL corrected XO rel time: 0.3802043
   Max SL corrected XO rel time: 42.2093889
   TM values from -50 to 38 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-05-18T07:58:11.380.062.750 (XO: 0.3802043, t=34.0°C)  Rank: 00009
   TM values: Max =    32  Min =   -41  Ave =    -4.27  Std =    11.35
   Min TM value:-41 at an offset of 125.538 usec -> 2017-05-18T07:58:11.380.188.287 (XO: 0.3803299)
   Max TM value:32 at an offset of 130.763 usec -> 2017-05-18T07:58:11.380.193.512 (XO: 0.3803351)
   Max dTM/dt [1/us]::2.85e+02 at an offset of 94.125 usec -> 2017-05-18T07:58:11.380.156.875 (XO: 0.3802985)
   Min dTM/dt [1/us]::-2.96e+02 at an offset of 150.776 usec -> 2017-05-18T07:58:11.380.213.525 (XO: 0.3803551)

 - Block number: 2 starting at 2017-05-18T07:58:40.319.612.750 (XO: 29.3197546, t=34.0°C)  Rank: 00009
   TM values: Max =    26  Min =   -35  Ave =    -4.23  Std =     9.09
   Min TM value:-35 at an offset of 147.251 usec -> 2017-05-18T07:58:40.319.760.000 (XO: 29.3199019)
   Max TM value:26 at an offset of 134.613 usec -> 2017-05-18T07:58:40.319.747.363 (XO: 29.3198892)
   Max dTM/dt [1/us]::2.30e+02 at an offset of 140.276 usec -> 2017-05-18T07:58:40.319.753.025 (XO: 29.3198949)
   Min dTM/dt [1/us]::-2.34e+02 at an offset of 131.701 usec -> 2017-05-18T07:58:40.319.744.450 (XO: 29.3198863)

 - Block number: 3 starting at 2017-05-18T07:58:53.208.958.750 (XO: 42.2090999, t=34.0°C)  Rank: 00018
   TM values: Max =    38  Min =   -50  Ave =    -4.34  Std =    15.66
   Min TM value:-50 at an offset of 124.838 usec -> 2017-05-18T07:58:53.209.083.587 (XO: 42.2092247)
   Max TM value:38 at an offset of 117.975 usec -> 2017-05-18T07:58:53.209.076.725 (XO: 42.2092179)
   Max dTM/dt [1/us]::3.48e+02 at an offset of 143.213 usec -> 2017-05-18T07:58:53.209.101.963 (XO: 42.2092431)
   Min dTM/dt [1/us]::-3.32e+02 at an offset of 109.288 usec -> 2017-05-18T07:58:53.209.068.037 (XO: 42.2092092)
