-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170512_013250_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.008587 to 35.897220
-each separately from 0s to 35s and from 8587us to 897220us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.059336 in 1 packets - #/No: 18/18
      20.828430 in 1 packets - #/No: 36/18
MIU packet time from 2017-05-12T01:32:53.008.587.000 to 2017-05-12T01:33:28.897.220.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.059336 in 1 packets - #/No: 18/18
      20.828430 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4469526 to 36.4468201
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999432 in 1 packets - #/No: 18/18
      19.9999243 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170512_013250_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4555409 to 1.4558299
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0592790  -- OK for SL --
   MIU time from 2017-05-12T01:32:53.008.587.000 to 2017-05-12T01:32:53.008.876.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.059336

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5151089 to 16.5153979
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.8283510  -- OK for SL --
   MIU time from 2017-05-12T01:33:08.068.212.000 to 2017-05-12T01:33:08.068.501.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.828430

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.3437489 to 37.3440379
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-05-12T01:33:28.896.931.000 to 2017-05-12T01:33:28.897.220.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-05-12T01:32:53.008.446.750
   Max SL corrected packet time: 2017-05-12T01:33:28.897.079.750
   Min SL corrected XO rel time: 0.0085883
   Max SL corrected XO rel time: 35.8972212
   TM values from -23 to 12 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-05-12T01:32:53.008.446.750 (XO: 0.0085883, t=34.0°C)  Rank: 00002
   TM values: Max =    12  Min =   -20  Ave =    -4.41  Std =     4.45
   Min TM value:-20 at an offset of 12.050 usec -> 2017-05-12T01:32:53.008.458.800 (XO: 0.0086004)
   Max TM value:12 at an offset of 121.088 usec -> 2017-05-12T01:32:53.008.567.837 (XO: 0.0087094)
   Max dTM/dt [1/us]::2.57e+02 at an offset of 89.300 usec -> 2017-05-12T01:32:53.008.536.050 (XO: 0.0086776)
   Min dTM/dt [1/us]::-2.19e+02 at an offset of 89.363 usec -> 2017-05-12T01:32:53.008.536.112 (XO: 0.0086777)

 - Block number: 2 starting at 2017-05-12T01:33:08.068.071.750 (XO: 15.0682134, t=34.0°C)  Rank: 00002
   TM values: Max =    12  Min =   -20  Ave =    -4.46  Std =     3.77
   Min TM value:-20 at an offset of 86.825 usec -> 2017-05-12T01:33:08.068.158.575 (XO: 15.0683002)
   Max TM value:12 at an offset of 33.163 usec -> 2017-05-12T01:33:08.068.104.912 (XO: 15.0682465)
   Max dTM/dt [1/us]::1.63e+02 at an offset of 273.089 usec -> 2017-05-12T01:33:08.068.344.838 (XO: 15.0684864)
   Min dTM/dt [1/us]::-1.80e+02 at an offset of 172.401 usec -> 2017-05-12T01:33:08.068.244.150 (XO: 15.0683858)

 - Block number: 3 starting at 2017-05-12T01:33:28.896.790.750 (XO: 35.8969322, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -23  Ave =    -4.35  Std =     3.89
   Min TM value:-23 at an offset of 180.151 usec -> 2017-05-12T01:33:28.896.970.900 (XO: 35.8971124)
   Max TM value:11 at an offset of 94.700 usec -> 2017-05-12T01:33:28.896.885.450 (XO: 35.8970269)
   Max dTM/dt [1/us]::1.96e+02 at an offset of 88.425 usec -> 2017-05-12T01:33:28.896.879.175 (XO: 35.8970206)
   Min dTM/dt [1/us]::-1.79e+02 at an offset of 179.863 usec -> 2017-05-12T01:33:28.896.970.613 (XO: 35.8971121)
