-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170506_120236_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 34 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 1 packets - #/No: 18/18
Total number of packets: 
      18 in 36 packets (all packets)
Hz and MHz counters from 5.016200 to 15.147984
-each separately from 5s to 15s and from 16200us to 147984us.
   Differences in neigboring packets: 
      0.000017 in 34 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.131206 in 1 packets - #/No: 18/18
MIU packet time from 2017-05-06T12:02:44.016.200.000 to 2017-05-06T12:02:54.147.984.000
   Differences in neigboring packets: 
      0.000017 in 34 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.131206 in 1 packets - #/No: 18/18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4659248 to 16.4658869
   Differences in neigboring packets: 
      0.0000000 in 34 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999621 in 1 packets - #/No: 18/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 36 packets (all packets)
Number of external events: 
      0 in 1 packets - #/No: 1/1
      1 in 35 packets - #/No: 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, 19/1, ...
Number of generated alerts: 
      0 in 1 packets - #/No: 1/1
      1 in 35 packets - #/No: 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, 19/1, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
Zero: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20170506_120236_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      96 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.4821261 to 6.4824151
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.1311670  -- OK for SL --
   MIU time from 2017-05-06T12:02:44.016.200.000 to 2017-05-06T12:02:44.016.489.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.131206

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      55 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6135821 to 16.6138711
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-05-06T12:02:54.147.695.000 to 2017-05-06T12:02:54.147.984.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-05-06T12:02:44.016.059.750
   Max SL corrected packet time: 2017-05-06T12:02:54.147.843.750
   Min SL corrected XO rel time: 5.0162014
   Max SL corrected XO rel time: 15.1479848
   TM values from -475 to 430 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-05-06T12:02:44.016.059.750 (XO: 5.0162014, t=34.0°C)  Rank: 00096
   TM values: Max =   414  Min =  -475  Ave =    -4.49  Std =    65.96
   Min TM value:-475 at an offset of 169.951 usec -> 2017-05-06T12:02:44.016.229.700 (XO: 5.0163713)
   Max TM value:414 at an offset of 113.663 usec -> 2017-05-06T12:02:44.016.173.412 (XO: 5.0163150)
   Max dTM/dt [1/us]::4.98e+03 at an offset of 113.563 usec -> 2017-05-06T12:02:44.016.173.312 (XO: 5.0163149)
   Min dTM/dt [1/us]::-3.68e+03 at an offset of 229.376 usec -> 2017-05-06T12:02:44.016.289.125 (XO: 5.0164307)

 - Block number: 2 starting at 2017-05-06T12:02:54.147.554.750 (XO: 15.1476958, t=34.0°C)  Rank: 00055
   TM values: Max =   430  Min =  -226  Ave =    -4.32  Std =    31.79
   Min TM value:-226 at an offset of 92.225 usec -> 2017-05-06T12:02:54.147.646.975 (XO: 15.1477880)
   Max TM value:430 at an offset of 121.763 usec -> 2017-05-06T12:02:54.147.676.512 (XO: 15.1478175)
   Max dTM/dt [1/us]::3.28e+03 at an offset of 95.325 usec -> 2017-05-06T12:02:54.147.650.075 (XO: 15.1477911)
   Min dTM/dt [1/us]::-2.95e+03 at an offset of 95.450 usec -> 2017-05-06T12:02:54.147.650.200 (XO: 15.1477912)
