-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170501_080909_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.003596 to 31.863380
-each separately from 0s to 31s and from 3596us to 863380us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.017719 in 1 packets - #/No: 18/18
      16.841198 in 1 packets - #/No: 36/18
MIU packet time from 2017-05-01T08:09:12.003.596.000 to 2017-05-01T08:09:43.863.380.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.017719 in 1 packets - #/No: 18/18
      16.841198 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4343100 to 32.4341933
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999435 in 1 packets - #/No: 18/18
      15.9999398 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170501_080909_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4379067 to 1.4381957
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0176630  -- OK for SL --
   MIU time from 2017-05-01T08:09:12.003.596.000 to 2017-05-01T08:09:12.003.885.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.017719

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4558587 to 16.4561477
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.8411350  -- OK for SL --
   MIU time from 2017-05-01T08:09:27.021.604.000 to 2017-05-01T08:09:27.021.893.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.841198

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.2972827 to 33.2975717
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-05-01T08:09:43.863.091.000 to 2017-05-01T08:09:43.863.380.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-05-01T08:09:12.003.455.750
   Max SL corrected packet time: 2017-05-01T08:09:43.863.239.750
   Min SL corrected XO rel time: 0.0035967
   Max SL corrected XO rel time: 31.8633816
   TM values from -20 to 12 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-05-01T08:09:12.003.455.750 (XO: 0.0035967, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -20  Ave =    -4.36  Std =     4.48
   Min TM value:-20 at an offset of 111.738 usec -> 2017-05-01T08:09:12.003.567.487 (XO: 0.0037085)
   Max TM value:11 at an offset of 139.788 usec -> 2017-05-01T08:09:12.003.595.538 (XO: 0.0037365)
   Max dTM/dt [1/us]::1.00e+02 at an offset of 195.663 usec -> 2017-05-01T08:09:12.003.651.413 (XO: 0.0037924)
   Min dTM/dt [1/us]::-1.14e+02 at an offset of 206.863 usec -> 2017-05-01T08:09:12.003.662.613 (XO: 0.0038036)

 - Block number: 2 starting at 2017-05-01T08:09:27.021.463.750 (XO: 15.0216053, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -19  Ave =    -4.47  Std =     4.67
   Min TM value:-19 at an offset of 129.025 usec -> 2017-05-01T08:09:27.021.592.775 (XO: 15.0217343)
   Max TM value:12 at an offset of 105.375 usec -> 2017-05-01T08:09:27.021.569.125 (XO: 15.0217107)
   Max dTM/dt [1/us]::9.50e+01 at an offset of 120.325 usec -> 2017-05-01T08:09:27.021.584.075 (XO: 15.0217256)
   Min dTM/dt [1/us]::-1.02e+02 at an offset of 218.426 usec -> 2017-05-01T08:09:27.021.682.175 (XO: 15.0218237)

 - Block number: 3 starting at 2017-05-01T08:09:43.862.950.750 (XO: 31.8630926, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -18  Ave =    -4.49  Std =     4.52
   Min TM value:-18 at an offset of 194.226 usec -> 2017-05-01T08:09:43.863.144.975 (XO: 31.8632868)
   Max TM value:10 at an offset of 135.551 usec -> 2017-05-01T08:09:43.863.086.300 (XO: 31.8632281)
   Max dTM/dt [1/us]::1.02e+02 at an offset of 127.688 usec -> 2017-05-01T08:09:43.863.078.437 (XO: 31.8632203)
   Min dTM/dt [1/us]::-9.30e+01 at an offset of 171.101 usec -> 2017-05-01T08:09:43.863.121.850 (XO: 31.8632637)
