-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170429_061245_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.991596 to 44.665030
-each separately from 3s to 44s and from 70326us to 991885us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.078441 in 1 packets - #/No: 18/18
      29.594126 in 1 packets - #/No: 36/18
MIU packet time from 2017-04-29T06:12:51.991.596.000 to 2017-04-29T06:13:32.665.030.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.078441 in 1 packets - #/No: 18/18
      29.594126 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4411103 to 45.4409560
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999548 in 1 packets - #/No: 18/18
      28.9998909 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170429_061245_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.4327044 to 5.4329934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0783990  -- OK for SL --
   MIU time from 2017-04-29T06:12:51.991.596.000 to 2017-04-29T06:12:51.991.885.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.078441

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5113924 to 16.5116814
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.5940150  -- OK for SL --
   MIU time from 2017-04-29T06:13:03.070.326.000 to 2017-04-29T06:13:03.070.615.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.594126

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.1056964 to 46.1059854
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-04-29T06:13:32.664.741.000 to 2017-04-29T06:13:32.665.030.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-04-29T06:12:51.991.455.750
   Max SL corrected packet time: 2017-04-29T06:13:32.664.889.750
   Min SL corrected XO rel time: 3.9915979
   Max SL corrected XO rel time: 44.6650335
   TM values from -28 to 21 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-04-29T06:12:51.991.455.750 (XO: 3.9915979, t=35.6°C)  Rank: 00004
   TM values: Max =    21  Min =   -25  Ave =    -4.64  Std =     6.32
   Min TM value:-25 at an offset of 229.513 usec -> 2017-04-29T06:12:51.991.685.263 (XO: 3.9918274)
   Max TM value:21 at an offset of 175.751 usec -> 2017-04-29T06:12:51.991.631.500 (XO: 3.9917736)
   Max dTM/dt [1/us]::2.33e+02 at an offset of 176.788 usec -> 2017-04-29T06:12:51.991.632.538 (XO: 3.9917747)
   Min dTM/dt [1/us]::-1.95e+02 at an offset of 118.150 usec -> 2017-04-29T06:12:51.991.573.900 (XO: 3.9917160)

 - Block number: 2 starting at 2017-04-29T06:13:03.070.185.750 (XO: 15.0703276, t=35.6°C)  Rank: 00003
   TM values: Max =    18  Min =   -25  Ave =    -4.56  Std =     6.27
   Min TM value:-25 at an offset of 204.513 usec -> 2017-04-29T06:13:03.070.390.263 (XO: 15.0705321)
   Max TM value:18 at an offset of 204.051 usec -> 2017-04-29T06:13:03.070.389.800 (XO: 15.0705316)
   Max dTM/dt [1/us]::1.90e+02 at an offset of 250.926 usec -> 2017-04-29T06:13:03.070.436.675 (XO: 15.0705785)
   Min dTM/dt [1/us]::-2.26e+02 at an offset of 100.888 usec -> 2017-04-29T06:13:03.070.286.637 (XO: 15.0704284)

 - Block number: 3 starting at 2017-04-29T06:13:32.664.600.750 (XO: 44.6647445, t=35.6°C)  Rank: 00004
   TM values: Max =    20  Min =   -28  Ave =    -4.54  Std =     6.66
   Min TM value:-28 at an offset of 63.363 usec -> 2017-04-29T06:13:32.664.664.112 (XO: 44.6648079)
   Max TM value:20 at an offset of 166.088 usec -> 2017-04-29T06:13:32.664.766.838 (XO: 44.6649106)
   Max dTM/dt [1/us]::2.04e+02 at an offset of 191.838 usec -> 2017-04-29T06:13:32.664.792.588 (XO: 44.6649364)
   Min dTM/dt [1/us]::-2.16e+02 at an offset of 126.838 usec -> 2017-04-29T06:13:32.664.727.587 (XO: 44.6648714)
